-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 12:47:28 2021 UTC
-- File 20210802_025625_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.557561 to 28.848093
-each separately from 8s to 28s and from 557561us to 848093us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.289954 in 1 packets: 18
MIU packet time from 2021-08-02T02:56:36.557.561.000 to 2021-08-02T02:56:56.848.093.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.289954 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9596355 to 29.9595689
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999334 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_025625_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.5171972 to 10.5174862
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.2898870  -- OK for SL --
   MIU time from 2021-08-02T02:56:36.557.561.000 to 2021-08-02T02:56:36.557.850.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.289954

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.8073732 to 30.8076622
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T02:56:56.847.804.000 to 2021-08-02T02:56:56.848.093.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.517197200 corrected by 40bit 10MHz counter of 9.959635500 obtained at 1Hz counter of 8. Result: 8.557563540
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T02:56:36.557.420.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T02:56:56.847.952.725
   Min 1st sample XO time among all SL packets: 8.557423264
   Max 1st sample XO time among all SL packets: 28.847955908
   TM values from -253 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T02:56:36.557.420.725 (XO: 8.557423264 - 8.557729266, t=37.2 degC)  Rank: 00080
   TM values: Max =   216  Min =  -253  Ave =    -4.47  Std =    22.02
   Min TM value:-253 at an offset of 93.888 usec -> 2021-08-02T02:56:36.557.514.612 (XO: 8.557517152)
   Max TM value:216 at an offset of 93.563 usec -> 2021-08-02T02:56:36.557.514.287 (XO: 8.557516827)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 96.250 usec -> 2021-08-02T02:56:36.557.516.975 (XO: 8.557519515)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 93.675 usec -> 2021-08-02T02:56:36.557.514.400 (XO: 8.557516940)

 - Block number: 2 starting at 2021-08-02T02:56:56.847.663.725 (XO: 28.847666907 - 28.847972908, t=37.2 degC)  Rank: 00081
   TM values: Max =   233  Min =  -245  Ave =    -4.45  Std =    18.22
   Min TM value:-245 at an offset of 119.188 usec -> 2021-08-02T02:56:56.847.782.912 (XO: 28.847786095)
   Max TM value:233 at an offset of 121.725 usec -> 2021-08-02T02:56:56.847.785.450 (XO: 28.847788632)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 121.563 usec -> 2021-08-02T02:56:56.847.785.287 (XO: 28.847788470)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 119.025 usec -> 2021-08-02T02:56:56.847.782.750 (XO: 28.847785932)
