-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 10:57:08 2021 UTC
-- File 20210802_012430_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 15.177025
-each separately from 0s to 15s and from 0us to 841253us.
   Differences in neigboring packets: 
      1.833939 in 1 packets: 1
      0.833944 in 2 packets: 2, 10
      0.833943 in 12 packets: 3, 4, 5, 6, 8, 9, 11, 12, 14, 15, 16, 17
      0.833941 in 2 packets: 7, 13
MIU packet time from 2021-08-02T01:24:33.000.000.000 to 2021-08-02T01:24:48.177.025.000
   Differences in neigboring packets: 
      1.833939 in 1 packets: 1
      0.833944 in 2 packets: 2, 10
      0.833943 in 12 packets: 3, 4, 5, 6, 8, 9, 11, 12, 14, 15, 16, 17
      0.833941 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 0.8853241 to 15.8852744
   Differences in neigboring packets: 
      0.9999968 in 2 packets: 1, 9
      0.9999966 in 4 packets: 2, 8, 11, 16
      0.9999967 in 9 packets: 3, 4, 5, 6, 10, 12, 14, 15, 17
      0.0000000 in 2 packets: 7, 13
   XO PPM: 
      -3.2 in 2 packets: 1, 9
      -3.4 in 4 packets: 2, 8, 11, 16
      -3.3 in 9 packets: 3, 4, 5, 6, 10, 12, 14, 15, 17
Temperature FPGA: 
      05 -> 48.5 degC in 5 packets: 1, 4, 5, 16, 18
      04 -> 49.8 degC in 10 packets: 2, 6, 7, 8, 9, 10, 11, 12, 15, 17
      06 -> 47.2 degC in 3 packets: 3, 13, 14
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 3 packets: 1, 2, 3
      1 in 7 packets: 4, 5, 6, 7, 8, 9, 10
      2 in 8 packets: 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 3 packets: 1, 2, 3
      1 in 7 packets: 4, 5, 6, 7, 8, 9, 10
      2 in 8 packets: 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_012430_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8853209 to 16.0623009
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T01:24:33.000.000.000 to 2021-08-02T01:24:48.177.025.000
      Difference between neigboring packets: 
      1.833939 in 1 packets: 1
      0.833944 in 2 packets: 2, 10
      0.833943 in 12 packets: 3, 4, 5, 6, 8, 9, 11, 12, 14, 15, 16, 17
      0.833941 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 15 XO calibration points at 1Hz counter from 0 to 14
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.885320900 corrected by 40bit 10MHz counter of 0.885324100 obtained at 1Hz counter of 0. Result: 1.000000000
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SR packets: 2021-08-02T01:24:32.999.999.925
   Max 1st sample MIU time among all SR packets: 2021-08-02T01:24:48.177.024.925
   Min 1st sample XO time among all SR packets: 0.999999925
   Max 1st sample XO time among all SR packets: 15.177027009
   TM values from -28 to 19 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2021-08-02T01:24:32.999.999.925 (XO: 0.999999925 - 1.000016925, t=37.2 degC)
   TM values: Max =    14  Min =   -28  Ave =    -4.33  Std =     6.06

 - Packet number: 2 starting at 2021-08-02T01:24:34.833.938.925 (XO: 1.833942760 - 1.833959760, t=37.2 degC)
   TM values: Max =    10  Min =   -22  Ave =    -4.21  Std =     5.36

 - Packet number: 3 starting at 2021-08-02T01:24:35.667.882.925 (XO: 2.667885529 - 2.667902529, t=37.2 degC)
   TM values: Max =    12  Min =   -26  Ave =    -4.34  Std =     6.05

 - Packet number: 4 starting at 2021-08-02T01:24:36.501.825.925 (XO: 3.501828281 - 3.501845281, t=37.2 degC)
   TM values: Max =    14  Min =   -27  Ave =    -4.66  Std =     6.34

 - Packet number: 5 starting at 2021-08-02T01:24:37.335.768.925 (XO: 4.335771033 - 4.335788033, t=37.2 degC)
   TM values: Max =    12  Min =   -20  Ave =    -4.24  Std =     5.30

 - Packet number: 6 starting at 2021-08-02T01:24:38.169.711.925 (XO: 5.169713785 - 5.169730785, t=37.2 degC)
   TM values: Max =    10  Min =   -20  Ave =    -4.60  Std =     5.24

 - Packet number: 7 starting at 2021-08-02T01:24:39.003.654.925 (XO: 6.003656537 - 6.003673537, t=37.2 degC)
   TM values: Max =    11  Min =   -18  Ave =    -4.46  Std =     4.55

 - Packet number: 8 starting at 2021-08-02T01:24:39.837.595.925 (XO: 6.837599373 - 6.837616373, t=37.2 degC)
   TM values: Max =    12  Min =   -20  Ave =    -4.47  Std =     5.25

 - Packet number: 9 starting at 2021-08-02T01:24:40.671.538.925 (XO: 7.671542074 - 7.671559074, t=37.2 degC)
   TM values: Max =    13  Min =   -22  Ave =    -4.10  Std =     6.32

 - Packet number: 10 starting at 2021-08-02T01:24:41.505.481.925 (XO: 8.505484793 - 8.505501793, t=37.2 degC)
   TM values: Max =    10  Min =   -19  Ave =    -4.04  Std =     5.14

 - Packet number: 11 starting at 2021-08-02T01:24:42.339.425.925 (XO: 9.339427579 - 9.339444579, t=37.2 degC)
   TM values: Max =    12  Min =   -22  Ave =    -4.54  Std =     6.02

 - Packet number: 12 starting at 2021-08-02T01:24:43.173.368.925 (XO: 10.173370397 - 10.173387397, t=37.2 degC)
   TM values: Max =    10  Min =   -21  Ave =    -4.59  Std =     5.08

 - Packet number: 13 starting at 2021-08-02T01:24:44.007.311.925 (XO: 11.007313149 - 11.007330149, t=37.2 degC)
   TM values: Max =    12  Min =   -19  Ave =    -4.13  Std =     4.38

 - Packet number: 14 starting at 2021-08-02T01:24:44.841.252.925 (XO: 11.841255901 - 11.841272901, t=37.2 degC)
   TM values: Max =    13  Min =   -26  Ave =    -4.45  Std =     5.76

 - Packet number: 15 starting at 2021-08-02T01:24:45.675.195.925 (XO: 12.675198653 - 12.675215653, t=37.2 degC)
   TM values: Max =    10  Min =   -20  Ave =    -4.41  Std =     4.57

 - Packet number: 16 starting at 2021-08-02T01:24:46.509.138.925 (XO: 13.509141456 - 13.509158456, t=37.2 degC)
   TM values: Max =     9  Min =   -20  Ave =    -4.31  Std =     5.26

 - Packet number: 17 starting at 2021-08-02T01:24:47.343.081.925 (XO: 14.343084257 - 14.343101257, t=37.2 degC)
   TM values: Max =    11  Min =   -19  Ave =    -4.45  Std =     4.99

 - Packet number: 18 starting at 2021-08-02T01:24:48.177.024.925 (XO: 15.177027009 - 15.177044009, t=37.2 degC)
   TM values: Max =    19  Min =   -25  Ave =    -4.23  Std =     6.89
