-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 10:53:33 2021 UTC
-- File 20210802_012307_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.931875 to 7.932164
-each separately from 7s to 7s and from 931875us to 932164us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-08-02T01:23:17.931.875.000 to 2021-08-02T01:23:17.932.164.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9131286 to 8.9131286
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_012307_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8450042 to 9.8452932
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T01:23:17.931.875.000 to 2021-08-02T01:23:17.932.164.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.845004200 corrected by 40bit 10MHz counter of 8.913128600 obtained at 1Hz counter of 7. Result: 7.931878675
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T01:23:17.931.734.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T01:23:17.932.023.725
   Min 1st sample XO time among all SL packets: 7.931738400
   Max 1st sample XO time among all SL packets: 7.932027401
   TM values from -268 to 210 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T01:23:17.931.734.725 (XO: 7.931738400 - 7.932044401, t=37.2 degC)  Rank: 00080
   TM values: Max =   210  Min =  -268  Ave =    -4.44  Std =    22.99
   Min TM value:-268 at an offset of 112.263 usec -> 2021-08-02T01:23:17.931.846.987 (XO: 7.931850663)
   Max TM value:210 at an offset of 109.963 usec -> 2021-08-02T01:23:17.931.844.687 (XO: 7.931848363)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 109.825 usec -> 2021-08-02T01:23:17.931.844.550 (XO: 7.931848225)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 112.113 usec -> 2021-08-02T01:23:17.931.846.837 (XO: 7.931850513)
