-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Dec  3 10:34:55 2021 UTC
-- File 20210802_005934_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.353259 to 22.023378
-each separately from 1s to 22s and from 23089us to 353548us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.669541 in 1 packets: 18
MIU packet time from 2021-08-02T00:59:38.353.259.000 to 2021-08-02T00:59:59.023.378.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.669541 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8102984 to 23.8102288
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999304 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210802_005934_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.1635588 to 3.1638478
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.6694710  -- OK for SL --
   MIU time from 2021-08-02T00:59:38.353.259.000 to 2021-08-02T00:59:38.353.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.669541

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.8333188 to 23.8336078
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-02T00:59:59.023.089.000 to 2021-08-02T00:59:59.023.378.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.163558800 corrected by 40bit 10MHz counter of 2.810298400 obtained at 1Hz counter of 1. Result: 1.353261601
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-02T00:59:38.353.118.725
   Max 1st sample MIU time among all SL packets: 2021-08-02T00:59:59.023.237.725
   Min 1st sample XO time among all SL packets: 1.353121326
   Max 1st sample XO time among all SL packets: 22.023238802
   TM values from -270 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-02T00:59:38.353.118.725 (XO: 1.353121326 - 1.353427327, t=37.2 degC)  Rank: 00092
   TM values: Max =   235  Min =  -231  Ave =    -4.45  Std =    20.72
   Min TM value:-231 at an offset of 119.863 usec -> 2021-08-02T00:59:38.353.238.587 (XO: 1.353241189)
   Max TM value:235 at an offset of 122.575 usec -> 2021-08-02T00:59:38.353.241.300 (XO: 1.353243901)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 122.463 usec -> 2021-08-02T00:59:38.353.241.187 (XO: 1.353243789)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 119.713 usec -> 2021-08-02T00:59:38.353.238.437 (XO: 1.353241039)

 - Block number: 2 starting at 2021-08-02T00:59:59.022.948.725 (XO: 22.022949801 - 22.023255802, t=37.2 degC)  Rank: 00079
   TM values: Max =   208  Min =  -270  Ave =    -4.37  Std =    26.90
   Min TM value:-270 at an offset of 93.913 usec -> 2021-08-02T00:59:59.023.042.637 (XO: 22.023043714)
   Max TM value:208 at an offset of 120.350 usec -> 2021-08-02T00:59:59.023.069.075 (XO: 22.023070151)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 123.050 usec -> 2021-08-02T00:59:59.023.071.775 (XO: 22.023072851)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 108.525 usec -> 2021-08-02T00:59:59.023.057.250 (XO: 22.023058326)
