-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Dec  2 13:42:50 2021 UTC
-- File 20210801_083304_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.738996 to 29.250626
-each separately from 1s to 29s and from 250337us to 739285us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.511052 in 1 packets: 18
MIU packet time from 2021-08-01T08:33:08.738.996.000 to 2021-08-01T08:33:36.250.626.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.511052 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9023365 to 30.9022430
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999065 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210801_083304_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6413332 to 3.6416222
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.5109590  -- OK for SL --
   MIU time from 2021-08-01T08:33:08.738.996.000 to 2021-08-01T08:33:08.739.285.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.511052

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.1525812 to 31.1528702
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-08-01T08:33:36.250.337.000 to 2021-08-01T08:33:36.250.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.641333200 corrected by 40bit 10MHz counter of 2.902336500 obtained at 1Hz counter of 1. Result: 1.738999139
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-08-01T08:33:08.738.855.725
   Max 1st sample MIU time among all SL packets: 2021-08-01T08:33:36.250.485.725
   Min 1st sample XO time among all SL packets: 1.738858863
   Max 1st sample XO time among all SL packets: 29.250487752
   TM values from -261 to 203 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-08-01T08:33:08.738.855.725 (XO: 1.738858863 - 1.739164864, t=37.2 degC)  Rank: 00074
   TM values: Max =   200  Min =  -261  Ave =    -4.35  Std =    20.75
   Min TM value:-261 at an offset of 114.425 usec -> 2021-08-01T08:33:08.738.970.150 (XO: 1.738973289)
   Max TM value:200 at an offset of 114.125 usec -> 2021-08-01T08:33:08.738.969.850 (XO: 1.738972989)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 116.888 usec -> 2021-08-01T08:33:08.738.972.612 (XO: 1.738975751)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 114.300 usec -> 2021-08-01T08:33:08.738.970.025 (XO: 1.738973164)

 - Block number: 2 starting at 2021-08-01T08:33:36.250.196.725 (XO: 29.250198751 - 29.250504752, t=37.2 degC)  Rank: 00082
   TM values: Max =   203  Min =  -253  Ave =    -4.46  Std =    21.40
   Min TM value:-253 at an offset of 77.150 usec -> 2021-08-01T08:33:36.250.273.875 (XO: 29.250275901)
   Max TM value:203 at an offset of 92.350 usec -> 2021-08-01T08:33:36.250.289.075 (XO: 29.250291101)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 92.163 usec -> 2021-08-01T08:33:36.250.288.887 (XO: 29.250290913)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 89.563 usec -> 2021-08-01T08:33:36.250.286.287 (XO: 29.250288313)
