-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  1 09:49:45 2021 UTC
-- File 20210730_121146_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.019294 to 21.389630
-each separately from 7s to 21s and from 19294us to 389630us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.369758 in 1 packets: 18
MIU packet time from 2021-07-30T12:11:56.019.294.000 to 2021-07-30T12:12:10.389.630.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.369758 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9420054 to 22.9419584
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999530 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C1 -> 37.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210730_121146_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.9613002 to 8.9615892
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3697110  -- OK for SL --
   MIU time from 2021-07-30T12:11:56.019.294.000 to 2021-07-30T12:11:56.019.583.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.369758

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.3313002 to 23.3315892
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-30T12:12:10.389.341.000 to 2021-07-30T12:12:10.389.630.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.961300200 corrected by 40bit 10MHz counter of 8.942005400 obtained at 1Hz counter of 7. Result: 7.019294866
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-30T12:11:56.019.153.725
   Max 1st sample MIU time among all SL packets: 2021-07-30T12:12:10.389.489.725
   Min 1st sample XO time among all SL packets: 7.019154590
   Max 1st sample XO time among all SL packets: 21.389491810
   TM values from -252 to 210 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-30T12:11:56.019.153.725 (XO: 7.019154590 - 7.019460591, t=35.6 degC)  Rank: 00088
   TM values: Max =   210  Min =  -252  Ave =    -4.59  Std =    24.07
   Min TM value:-252 at an offset of 106.763 usec -> 2021-07-30T12:11:56.019.260.487 (XO: 7.019261353)
   Max TM value:210 at an offset of 80.725 usec -> 2021-07-30T12:11:56.019.234.450 (XO: 7.019235315)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 109.125 usec -> 2021-07-30T12:11:56.019.262.850 (XO: 7.019263715)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 106.575 usec -> 2021-07-30T12:11:56.019.260.300 (XO: 7.019261165)

 - Block number: 2 starting at 2021-07-30T12:12:10.389.200.725 (XO: 21.389202809 - 21.389508810, t=37.2 degC)  Rank: 00083
   TM values: Max =   165  Min =  -194  Ave =    -4.50  Std =    19.44
   Min TM value:-194 at an offset of 106.063 usec -> 2021-07-30T12:12:10.389.306.787 (XO: 21.389308872)
   Max TM value:165 at an offset of 108.850 usec -> 2021-07-30T12:12:10.389.309.575 (XO: 21.389311660)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 108.738 usec -> 2021-07-30T12:12:10.389.309.462 (XO: 21.389311547)
   Min dTM/dt [1/us]:-1.49e+03 at an offset of 92.963 usec -> 2021-07-30T12:12:10.389.293.687 (XO: 21.389295772)
