-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  1 07:47:46 2021 UTC
-- File 20210730_111448_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.315190 to 62.681439
-each separately from 2s to 62s and from 113748us to 711902us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.798269 in 1 packets: 18
      14.048350 in 1 packets: 36
      22.548937 in 1 packets: 54
      9.969248 in 1 packets: 72
MIU packet time from 2021-07-30T11:14:53.315.190.000 to 2021-07-30T11:15:53.681.439.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.798269 in 1 packets: 18
      14.048350 in 1 packets: 36
      22.548937 in 1 packets: 54
      9.969248 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9376154 to 63.9374152
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999533 in 1 packets: 18
      13.9999532 in 1 packets: 36
      21.9999267 in 1 packets: 54
      9.9999666 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210730_111448_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2528061 to 4.2530951
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7982230  -- OK for SL --
   MIU time from 2021-07-30T11:14:53.315.190.000 to 2021-07-30T11:14:53.315.479.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.798269

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0513181 to 18.0516071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0483030  -- OK for SL --
   MIU time from 2021-07-30T11:15:07.113.748.000 to 2021-07-30T11:15:07.114.037.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.048350

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0999101 to 32.1001991
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.5488630  -- OK for SL --
   MIU time from 2021-07-30T11:15:21.162.387.000 to 2021-07-30T11:15:21.162.676.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.548937

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.6490621 to 54.6493511
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9692150  -- OK for SL --
   MIU time from 2021-07-30T11:15:43.711.613.000 to 2021-07-30T11:15:43.711.902.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.969248

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.6185661 to 64.6188551
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-30T11:15:53.681.150.000 to 2021-07-30T11:15:53.681.439.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.252806100 corrected by 40bit 10MHz counter of 3.937615400 obtained at 1Hz counter of 2. Result: 2.315191772
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-30T11:14:53.315.049.725
   Max 1st sample MIU time among all SL packets: 2021-07-30T11:15:53.681.298.725
   Min 1st sample XO time among all SL packets: 2.315051496
   Max 1st sample XO time among all SL packets: 62.681301941
   TM values from -258 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-30T11:14:53.315.049.725 (XO: 2.315051496 - 2.315357497, t=35.6 degC)  Rank: 00091
   TM values: Max =   187  Min =  -243  Ave =    -4.52  Std =    20.71
   Min TM value:-243 at an offset of 124.775 usec -> 2021-07-30T11:14:53.315.174.500 (XO: 2.315176272)
   Max TM value:187 at an offset of 127.425 usec -> 2021-07-30T11:14:53.315.177.150 (XO: 2.315178922)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 127.250 usec -> 2021-07-30T11:14:53.315.176.975 (XO: 2.315178747)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 124.588 usec -> 2021-07-30T11:14:53.315.174.312 (XO: 2.315176084)

 - Block number: 2 starting at 2021-07-30T11:15:07.113.607.725 (XO: 16.113609511 - 16.113915512, t=35.6 degC)  Rank: 00080
   TM values: Max =   182  Min =  -258  Ave =    -4.42  Std =    20.46
   Min TM value:-258 at an offset of 113.300 usec -> 2021-07-30T11:15:07.113.721.025 (XO: 16.113722812)
   Max TM value:182 at an offset of 115.900 usec -> 2021-07-30T11:15:07.113.723.625 (XO: 16.113725412)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 115.725 usec -> 2021-07-30T11:15:07.113.723.450 (XO: 16.113725237)
   Min dTM/dt [1/us]:-1.51e+03 at an offset of 113.113 usec -> 2021-07-30T11:15:07.113.720.837 (XO: 16.113722624)

 - Block number: 3 starting at 2021-07-30T11:15:21.162.246.725 (XO: 30.162248460 - 30.162554461, t=35.6 degC)  Rank: 00092
   TM values: Max =   248  Min =  -249  Ave =    -4.50  Std =    20.04
   Min TM value:-249 at an offset of 128.650 usec -> 2021-07-30T11:15:21.162.375.375 (XO: 30.162377111)
   Max TM value:248 at an offset of 126.513 usec -> 2021-07-30T11:15:21.162.373.237 (XO: 30.162374973)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 126.400 usec -> 2021-07-30T11:15:21.162.373.125 (XO: 30.162374861)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 128.488 usec -> 2021-07-30T11:15:21.162.375.212 (XO: 30.162376948)

 - Block number: 4 starting at 2021-07-30T11:15:43.711.472.725 (XO: 52.711475644 - 52.711781645, t=35.6 degC)  Rank: 00078
   TM values: Max =   208  Min =  -252  Ave =    -4.44  Std =    17.57
   Min TM value:-252 at an offset of 127.413 usec -> 2021-07-30T11:15:43.711.600.137 (XO: 52.711603057)
   Max TM value:208 at an offset of 127.113 usec -> 2021-07-30T11:15:43.711.599.837 (XO: 52.711602757)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 129.813 usec -> 2021-07-30T11:15:43.711.602.537 (XO: 52.711605457)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 127.213 usec -> 2021-07-30T11:15:43.711.599.937 (XO: 52.711602857)

 - Block number: 5 starting at 2021-07-30T11:15:53.681.009.725 (XO: 62.681012940 - 62.681318941, t=35.6 degC)  Rank: 00079
   TM values: Max =   210  Min =  -247  Ave =    -4.43  Std =    20.30
   Min TM value:-247 at an offset of 115.988 usec -> 2021-07-30T11:15:53.681.125.712 (XO: 62.681128928)
   Max TM value:210 at an offset of 115.688 usec -> 2021-07-30T11:15:53.681.125.412 (XO: 62.681128628)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 113.563 usec -> 2021-07-30T11:15:53.681.123.287 (XO: 62.681126503)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 115.813 usec -> 2021-07-30T11:15:53.681.125.537 (XO: 62.681128753)
