-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Dec  1 02:31:15 2021 UTC
-- File 20210730_082154_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.015715 to 17.045906
-each separately from 13s to 17s and from 15715us to 45906us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.029613 in 1 packets: 18
MIU packet time from 2021-07-30T08:22:10.015.715.000 to 2021-07-30T08:22:14.045.906.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.029613 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8738303 to 18.8738170
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999867 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210730_082154_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8895467 to 14.8898357
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0295990  -- OK for SL --
   MIU time from 2021-07-30T08:22:10.015.715.000 to 2021-07-30T08:22:10.016.004.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.029613

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9194347 to 18.9197237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-30T08:22:14.045.617.000 to 2021-07-30T08:22:14.045.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.889546700 corrected by 40bit 10MHz counter of 14.873830300 obtained at 1Hz counter of 13. Result: 13.015716452
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-30T08:22:10.015.574.725
   Max 1st sample MIU time among all SL packets: 2021-07-30T08:22:14.045.765.725
   Min 1st sample XO time among all SL packets: 13.015576176
   Max 1st sample XO time among all SL packets: 17.045766576
   TM values from -224 to 197 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-30T08:22:10.015.574.725 (XO: 13.015576176 - 13.015882177, t=37.2 degC)  Rank: 00084
   TM values: Max =   197  Min =  -224  Ave =    -4.58  Std =    16.50
   Min TM value:-224 at an offset of 113.913 usec -> 2021-07-30T08:22:10.015.688.637 (XO: 13.015690089)
   Max TM value:197 at an offset of 116.488 usec -> 2021-07-30T08:22:10.015.691.212 (XO: 13.015692664)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 116.300 usec -> 2021-07-30T08:22:10.015.691.025 (XO: 13.015692477)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 113.788 usec -> 2021-07-30T08:22:10.015.688.512 (XO: 13.015689964)

 - Block number: 2 starting at 2021-07-30T08:22:14.045.476.725 (XO: 17.045477575 - 17.045783576, t=37.2 degC)  Rank: 00085
   TM values: Max =   183  Min =  -217  Ave =    -4.45  Std =    16.73
   Min TM value:-217 at an offset of 124.038 usec -> 2021-07-30T08:22:14.045.600.762 (XO: 17.045601613)
   Max TM value:183 at an offset of 126.600 usec -> 2021-07-30T08:22:14.045.603.325 (XO: 17.045604175)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 126.425 usec -> 2021-07-30T08:22:14.045.603.150 (XO: 17.045604000)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 123.838 usec -> 2021-07-30T08:22:14.045.600.562 (XO: 17.045601413)
