-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 30 11:37:02 2021 UTC
-- File 20210729_120007_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.452222 to 15.390720
-each separately from 10s to 15s and from 390431us to 452511us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.937920 in 1 packets: 18
MIU packet time from 2021-07-29T12:00:20.452.222.000 to 2021-07-29T12:00:25.390.720.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.937920 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9164466 to 16.9164299
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999833 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210729_120007_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.3686695 to 12.3689585
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.9379030  -- OK for SL --
   MIU time from 2021-07-29T12:00:20.452.222.000 to 2021-07-29T12:00:20.452.511.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.937920

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3068615 to 17.3071505
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-29T12:00:25.390.431.000 to 2021-07-29T12:00:25.390.720.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.368669500 corrected by 40bit 10MHz counter of 11.916446600 obtained at 1Hz counter of 10. Result: 10.452224392
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-29T12:00:20.452.081.725
   Max 1st sample MIU time among all SL packets: 2021-07-29T12:00:25.390.579.725
   Min 1st sample XO time among all SL packets: 10.452084117
   Max 1st sample XO time among all SL packets: 15.390581614
   TM values from -251 to 203 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-29T12:00:20.452.081.725 (XO: 10.452084117 - 10.452390118, t=37.2 degC)  Rank: 00084
   TM values: Max =   203  Min =  -251  Ave =    -4.61  Std =    22.52
   Min TM value:-251 at an offset of 126.750 usec -> 2021-07-29T12:00:20.452.208.475 (XO: 10.452210867)
   Max TM value:203 at an offset of 126.463 usec -> 2021-07-29T12:00:20.452.208.187 (XO: 10.452210580)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 129.125 usec -> 2021-07-29T12:00:20.452.210.850 (XO: 10.452213242)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 126.575 usec -> 2021-07-29T12:00:20.452.208.300 (XO: 10.452210692)

 - Block number: 2 starting at 2021-07-29T12:00:25.390.290.725 (XO: 15.390292613 - 15.390598614, t=37.2 degC)  Rank: 00077
   TM values: Max =   181  Min =  -215  Ave =    -4.59  Std =    18.35
   Min TM value:-215 at an offset of 114.513 usec -> 2021-07-29T12:00:25.390.405.237 (XO: 15.390407126)
   Max TM value:181 at an offset of 117.163 usec -> 2021-07-29T12:00:25.390.407.887 (XO: 15.390409776)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 117.000 usec -> 2021-07-29T12:00:25.390.407.725 (XO: 15.390409613)
   Min dTM/dt [1/us]:-1.69e+03 at an offset of 114.300 usec -> 2021-07-29T12:00:25.390.405.025 (XO: 15.390406913)
