-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 26 15:25:46 2021 UTC
-- File 20210725_171007_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 15.177026
-each separately from 0s to 15s and from 0us to 841253us.
   Differences in neigboring packets: 
      1.833940 in 1 packets: 1
      0.833943 in 10 packets: 2, 3, 4, 6, 9, 10, 11, 15, 16, 17
      0.833944 in 4 packets: 5, 8, 12, 14
      0.833940 in 2 packets: 7, 13
MIU packet time from 2021-07-25T17:10:10.000.000.000 to 2021-07-25T17:10:25.177.026.000
   Differences in neigboring packets: 
      1.833940 in 1 packets: 1
      0.833943 in 10 packets: 2, 3, 4, 6, 9, 10, 11, 15, 16, 17
      0.833944 in 4 packets: 5, 8, 12, 14
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 0.8497960 to 15.8497455
   Differences in neigboring packets: 
      0.9999967 in 7 packets: 1, 4, 6, 9, 11, 12, 17
      0.9999966 in 6 packets: 2, 3, 8, 14, 15, 16
      0.9999965 in 2 packets: 5, 10
      0.0000000 in 2 packets: 7, 13
   XO PPM: 
      -3.3 in 7 packets: 1, 4, 6, 9, 11, 12, 17
      -3.4 in 6 packets: 2, 3, 8, 14, 15, 16
      -3.5 in 2 packets: 5, 10
Temperature FPGA: 
      05 -> 48.5 degC in 17 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 1 packets: 10
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 8 packets: 1, 2, 3, 4, 5, 6, 7, 8
      1 in 4 packets: 9, 10, 11, 12
      2 in 6 packets: 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 8 packets: 1, 2, 3, 4, 5, 6, 7, 8
      1 in 4 packets: 9, 10, 11, 12
      2 in 6 packets: 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210725_171007_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8497927 to 16.0267727
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-25T17:10:10.000.000.000 to 2021-07-25T17:10:25.177.026.000
      Difference between neigboring packets: 
      1.833940 in 1 packets: 1
      0.833943 in 10 packets: 2, 3, 4, 6, 9, 10, 11, 15, 16, 17
      0.833944 in 4 packets: 5, 8, 12, 14
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 15 XO calibration points at 1Hz counter from 0 to 14
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.849792700 corrected by 40bit 10MHz counter of 0.849796000 obtained at 1Hz counter of 0. Result: 1.000000000
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SR packets: 2021-07-25T17:10:09.999.999.925
   Max 1st sample MIU time among all SR packets: 2021-07-25T17:10:25.177.025.925
   Min 1st sample XO time among all SR packets: 0.999999925
   Max 1st sample XO time among all SR packets: 15.177027709
   TM values from -39 to 26 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2021-07-25T17:10:09.999.999.925 (XO: 0.999999925 - 1.000016925, t=37.2 degC)
   TM values: Max =    23  Min =   -34  Ave =    -4.58  Std =     9.73

 - Packet number: 2 starting at 2021-07-25T17:10:11.833.939.925 (XO: 1.833942760 - 1.833959760, t=37.2 degC)
   TM values: Max =    23  Min =   -34  Ave =    -4.31  Std =     9.34

 - Packet number: 3 starting at 2021-07-25T17:10:12.667.882.925 (XO: 2.667885596 - 2.667902596, t=37.2 degC)
   TM values: Max =    26  Min =   -39  Ave =    -4.56  Std =    12.08

 - Packet number: 4 starting at 2021-07-25T17:10:13.501.825.925 (XO: 3.501828381 - 3.501845381, t=37.2 degC)
   TM values: Max =    21  Min =   -30  Ave =    -4.47  Std =     8.91

 - Packet number: 5 starting at 2021-07-25T17:10:14.335.768.925 (XO: 4.335771200 - 4.335788200, t=37.2 degC)
   TM values: Max =    22  Min =   -36  Ave =    -4.49  Std =     9.07

 - Packet number: 6 starting at 2021-07-25T17:10:15.169.712.925 (XO: 5.169714085 - 5.169731085, t=37.2 degC)
   TM values: Max =    19  Min =   -30  Ave =    -4.57  Std =     7.99

 - Packet number: 7 starting at 2021-07-25T17:10:16.003.655.925 (XO: 6.003656837 - 6.003673837, t=37.2 degC)
   TM values: Max =    19  Min =   -29  Ave =    -4.48  Std =     7.03

 - Packet number: 8 starting at 2021-07-25T17:10:16.837.595.925 (XO: 6.837599673 - 6.837616673, t=37.2 degC)
   TM values: Max =    22  Min =   -27  Ave =    -4.54  Std =     7.53

 - Packet number: 9 starting at 2021-07-25T17:10:17.671.539.925 (XO: 7.671542441 - 7.671559441, t=37.2 degC)
   TM values: Max =    21  Min =   -32  Ave =    -4.43  Std =     9.77

 - Packet number: 10 starting at 2021-07-25T17:10:18.505.482.925 (XO: 8.505485294 - 8.505502294, t=37.2 degC)
   TM values: Max =    20  Min =   -27  Ave =    -4.28  Std =     8.34

 - Packet number: 11 starting at 2021-07-25T17:10:19.339.425.925 (XO: 9.339428145 - 9.339445145, t=37.2 degC)
   TM values: Max =    18  Min =   -26  Ave =    -4.53  Std =     7.33

 - Packet number: 12 starting at 2021-07-25T17:10:20.173.368.925 (XO: 10.173370897 - 10.173387897, t=37.2 degC)
   TM values: Max =    17  Min =   -22  Ave =    -4.41  Std =     5.52

 - Packet number: 13 starting at 2021-07-25T17:10:21.007.312.925 (XO: 11.007313650 - 11.007330650, t=37.2 degC)
   TM values: Max =    19  Min =   -25  Ave =    -4.41  Std =     7.82

 - Packet number: 14 starting at 2021-07-25T17:10:21.841.252.925 (XO: 11.841256485 - 11.841273485, t=37.2 degC)
   TM values: Max =    19  Min =   -28  Ave =    -4.25  Std =     6.71

 - Packet number: 15 starting at 2021-07-25T17:10:22.675.196.925 (XO: 12.675199321 - 12.675216321, t=37.2 degC)
   TM values: Max =    19  Min =   -28  Ave =    -4.60  Std =     7.74

 - Packet number: 16 starting at 2021-07-25T17:10:23.509.139.925 (XO: 13.509142156 - 13.509159156, t=37.2 degC)
   TM values: Max =    21  Min =   -28  Ave =    -4.41  Std =     7.93

 - Packet number: 17 starting at 2021-07-25T17:10:24.343.082.925 (XO: 14.343084957 - 14.343101957, t=37.2 degC)
   TM values: Max =    18  Min =   -30  Ave =    -4.53  Std =     8.11

 - Packet number: 18 starting at 2021-07-25T17:10:25.177.025.925 (XO: 15.177027709 - 15.177044709, t=37.2 degC)
   TM values: Max =    14  Min =   -24  Ave =    -4.46  Std =     6.32
