-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 24 04:28:29 2021 UTC
-- File 20210723_101502_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.441661 to 88.812288
-each separately from 6s to 88s and from 131868us to 872146us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.709861 in 1 packets: 18
      16.979768 in 1 packets: 36
      6.739700 in 1 packets: 54
      9.639680 in 1 packets: 72
      28.299884 in 1 packets: 90
MIU packet time from 2021-07-23T10:15:11.441.661.000 to 2021-07-23T10:16:33.812.288.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.709861 in 1 packets: 18
      16.979768 in 1 packets: 36
      6.739700 in 1 packets: 54
      9.639680 in 1 packets: 72
      28.299884 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9162272 to 89.9159531
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999296 in 1 packets: 18
      16.9999430 in 1 packets: 36
      5.9999800 in 1 packets: 54
      9.9999667 in 1 packets: 72
      27.9999066 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210723_101502_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.3578890 to 8.3581780
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7097910  -- OK for SL --
   MIU time from 2021-07-23T10:15:11.441.661.000 to 2021-07-23T10:15:11.441.950.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.709861

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0679690 to 29.0682580
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9797110  -- OK for SL --
   MIU time from 2021-07-23T10:15:32.151.811.000 to 2021-07-23T10:15:32.152.100.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.979768

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.0479690 to 46.0482580
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.7396790  -- OK for SL --
   MIU time from 2021-07-23T10:15:49.131.868.000 to 2021-07-23T10:15:49.132.157.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.739700

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.7879370 to 52.7882260
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.6396470  -- OK for SL --
   MIU time from 2021-07-23T10:15:55.871.857.000 to 2021-07-23T10:15:55.872.146.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.639680

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.4278730 to 62.4281620
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.2997910  -- OK for SL --
   MIU time from 2021-07-23T10:16:05.511.826.000 to 2021-07-23T10:16:05.512.115.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.299884

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 90.7279530 to 90.7282420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-23T10:16:33.811.999.000 to 2021-07-23T10:16:33.812.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.357889000 corrected by 40bit 10MHz counter of 7.916227200 obtained at 1Hz counter of 6. Result: 6.441663257
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-23T10:15:11.441.520.725
   Max 1st sample MIU time among all SL packets: 2021-07-23T10:16:33.812.147.725
   Min 1st sample XO time among all SL packets: 6.441522982
   Max 1st sample XO time among all SL packets: 88.812151386
   TM values from -244 to 233 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-23T10:15:11.441.520.725 (XO: 6.441522982 - 6.441828983, t=37.2 degC)  Rank: 00078
   TM values: Max =   233  Min =  -244  Ave =    -4.45  Std =    30.98
   Min TM value:-244 at an offset of 194.513 usec -> 2021-07-23T10:15:11.441.715.238 (XO: 6.441717495)
   Max TM value:233 at an offset of 192.201 usec -> 2021-07-23T10:15:11.441.712.925 (XO: 6.441715183)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 192.076 usec -> 2021-07-23T10:15:11.441.712.800 (XO: 6.441715058)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 208.438 usec -> 2021-07-23T10:15:11.441.729.163 (XO: 6.441731420)

 - Block number: 2 starting at 2021-07-23T10:15:32.151.670.725 (XO: 27.151672441 - 27.151978442, t=37.2 degC)  Rank: 00087
   TM values: Max =   202  Min =  -236  Ave =    -4.50  Std =    20.04
   Min TM value:-236 at an offset of 122.750 usec -> 2021-07-23T10:15:32.151.793.475 (XO: 27.151795191)
   Max TM value:202 at an offset of 122.463 usec -> 2021-07-23T10:15:32.151.793.187 (XO: 27.151794904)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 125.225 usec -> 2021-07-23T10:15:32.151.795.950 (XO: 27.151797666)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 122.575 usec -> 2021-07-23T10:15:32.151.793.300 (XO: 27.151795016)

 - Block number: 3 starting at 2021-07-23T10:15:49.131.727.725 (XO: 44.131729360 - 44.132035361, t=37.2 degC)  Rank: 00079
   TM values: Max =   208  Min =  -243  Ave =    -4.61  Std =    17.98
   Min TM value:-243 at an offset of 122.313 usec -> 2021-07-23T10:15:49.131.850.037 (XO: 44.131851673)
   Max TM value:208 at an offset of 122.013 usec -> 2021-07-23T10:15:49.131.849.737 (XO: 44.131851373)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 124.763 usec -> 2021-07-23T10:15:49.131.852.487 (XO: 44.131854123)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 122.125 usec -> 2021-07-23T10:15:49.131.849.850 (XO: 44.131851485)

 - Block number: 4 starting at 2021-07-23T10:15:55.871.716.725 (XO: 50.871719802 - 50.872025803, t=37.2 degC)  Rank: 00078
   TM values: Max =   202  Min =  -224  Ave =    -4.62  Std =    20.33
   Min TM value:-224 at an offset of 124.150 usec -> 2021-07-23T10:15:55.871.840.875 (XO: 50.871843952)
   Max TM value:202 at an offset of 126.863 usec -> 2021-07-23T10:15:55.871.843.587 (XO: 50.871846665)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 126.688 usec -> 2021-07-23T10:15:55.871.843.412 (XO: 50.871846490)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 123.963 usec -> 2021-07-23T10:15:55.871.840.687 (XO: 50.871843765)

 - Block number: 5 starting at 2021-07-23T10:16:05.511.685.725 (XO: 60.511687914 - 60.511993915, t=37.2 degC)  Rank: 00075
   TM values: Max =   203  Min =  -229  Ave =    -4.46  Std =    19.14
   Min TM value:-229 at an offset of 110.738 usec -> 2021-07-23T10:16:05.511.796.462 (XO: 60.511798651)
   Max TM value:203 at an offset of 113.463 usec -> 2021-07-23T10:16:05.511.799.187 (XO: 60.511801376)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 113.288 usec -> 2021-07-23T10:16:05.511.799.012 (XO: 60.511801201)
   Min dTM/dt [1/us]:-1.40e+03 at an offset of 98.975 usec -> 2021-07-23T10:16:05.511.784.700 (XO: 60.511786889)

 - Block number: 6 starting at 2021-07-23T10:16:33.811.858.725 (XO: 88.811862385 - 88.812168386, t=37.2 degC)  Rank: 00081
   TM values: Max =   216  Min =  -244  Ave =    -4.40  Std =    21.93
   Min TM value:-244 at an offset of 126.825 usec -> 2021-07-23T10:16:33.811.985.550 (XO: 88.811989211)
   Max TM value:216 at an offset of 124.250 usec -> 2021-07-23T10:16:33.811.982.975 (XO: 88.811986636)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 126.950 usec -> 2021-07-23T10:16:33.811.985.675 (XO: 88.811989336)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 124.375 usec -> 2021-07-23T10:16:33.811.983.100 (XO: 88.811986761)
