-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Nov 20 02:58:01 2021 UTC
-- File 20210721_095704_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.121383 to 20.111778
-each separately from 12s to 20s and from 111489us to 121672us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.989817 in 1 packets: 18
MIU packet time from 2021-07-21T09:57:19.121.383.000 to 2021-07-21T09:57:27.111.778.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.989817 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9049863 to 21.9049598
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999735 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210721_095704_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.0263700 to 14.0266590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9897910  -- OK for SL --
   MIU time from 2021-07-21T09:57:19.121.383.000 to 2021-07-21T09:57:19.121.672.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.989817

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.0164500 to 22.0167390
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-21T09:57:27.111.489.000 to 2021-07-21T09:57:27.111.778.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.026370000 corrected by 40bit 10MHz counter of 13.904986300 obtained at 1Hz counter of 12. Result: 12.121384101
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-21T09:57:19.121.242.725
   Max 1st sample MIU time among all SL packets: 2021-07-21T09:57:27.111.637.725
   Min 1st sample XO time among all SL packets: 12.121243825
   Max 1st sample XO time among all SL packets: 20.111639293
   TM values from -247 to 220 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-21T09:57:19.121.242.725 (XO: 12.121243825 - 12.121549826, t=37.2 degC)  Rank: 00074
   TM values: Max =   180  Min =  -214  Ave =    -4.44  Std =    20.82
   Min TM value:-214 at an offset of 117.738 usec -> 2021-07-21T09:57:19.121.360.462 (XO: 12.121361563)
   Max TM value:180 at an offset of 120.500 usec -> 2021-07-21T09:57:19.121.363.225 (XO: 12.121364326)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 120.325 usec -> 2021-07-21T09:57:19.121.363.050 (XO: 12.121364151)
   Min dTM/dt [1/us]:-1.52e+03 at an offset of 105.613 usec -> 2021-07-21T09:57:19.121.348.337 (XO: 12.121349438)

 - Block number: 2 starting at 2021-07-21T09:57:27.111.348.725 (XO: 20.111350292 - 20.111656293, t=37.2 degC)  Rank: 00085
   TM values: Max =   220  Min =  -247  Ave =    -4.51  Std =    22.84
   Min TM value:-247 at an offset of 106.563 usec -> 2021-07-21T09:57:27.111.455.287 (XO: 20.111456855)
   Max TM value:220 at an offset of 104.463 usec -> 2021-07-21T09:57:27.111.453.187 (XO: 20.111454755)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 104.325 usec -> 2021-07-21T09:57:27.111.453.050 (XO: 20.111454618)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 106.438 usec -> 2021-07-21T09:57:27.111.455.162 (XO: 20.111456730)
