-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 19 05:39:47 2021 UTC
-- File 20210720_090824_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.153128 to 67.535402
-each separately from 14s to 67s and from 153128us to 974512us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.569782 in 1 packets: 18
      24.250735 in 1 packets: 36
      15.390148 in 1 packets: 54
      11.170164 in 1 packets: 72
MIU packet time from 2021-07-20T09:08:41.153.128.000 to 2021-07-20T09:09:34.535.402.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.569782 in 1 packets: 18
      24.250735 in 1 packets: 36
      15.390148 in 1 packets: 54
      11.170164 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8369489 to 68.8367724
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.9999934 in 1 packets: 18
      23.9999202 in 1 packets: 36
      15.9999466 in 1 packets: 54
      10.9999633 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      3 in 1 packets: 1
      4 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      5 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      6 in 1 packets: 37
      7 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      8 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      9 in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      3 in 1 packets: 1
      4 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      5 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      6 in 1 packets: 37
      7 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      8 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      9 in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210720_090824_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.9900781 to 15.9903671
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.5697750  -- OK for SL --
   MIU time from 2021-07-20T09:08:41.153.128.000 to 2021-07-20T09:08:41.153.417.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.569782

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.5601421 to 18.5604311
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.2506550  -- OK for SL --
   MIU time from 2021-07-20T09:08:43.723.199.000 to 2021-07-20T09:08:43.723.488.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.250735

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.8110861 to 42.8113751
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3900950  -- OK for SL --
   MIU time from 2021-07-20T09:09:07.974.223.000 to 2021-07-20T09:09:07.974.512.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.390148

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.2014701 to 58.2017591
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1701270  -- OK for SL --
   MIU time from 2021-07-20T09:09:23.364.660.000 to 2021-07-20T09:09:23.364.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.170164

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.3718861 to 69.3721751
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-20T09:09:34.535.113.000 to 2021-07-20T09:09:34.535.402.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.990078100 corrected by 40bit 10MHz counter of 15.836948900 obtained at 1Hz counter of 14. Result: 14.153129705
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-20T09:08:41.152.987.725
   Max 1st sample MIU time among all SL packets: 2021-07-20T09:09:34.535.261.725
   Min 1st sample XO time among all SL packets: 14.152989430
   Max 1st sample XO time among all SL packets: 67.535264245
   TM values from -273 to 283 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-20T09:08:41.152.987.725 (XO: 14.152989430 - 14.153295431, t=37.2 degC)  Rank: 00086
   TM values: Max =   283  Min =  -269  Ave =    -4.63  Std =    24.77
   Min TM value:-269 at an offset of 105.625 usec -> 2021-07-20T09:08:41.153.093.350 (XO: 14.153095055)
   Max TM value:283 at an offset of 103.500 usec -> 2021-07-20T09:08:41.153.091.225 (XO: 14.153092930)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 103.375 usec -> 2021-07-20T09:08:41.153.091.100 (XO: 14.153092805)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 105.475 usec -> 2021-07-20T09:08:41.153.093.200 (XO: 14.153094905)

 - Block number: 2 starting at 2021-07-20T09:08:43.723.058.725 (XO: 16.723061983 - 16.723367984, t=37.2 degC)  Rank: 00082
   TM values: Max =   204  Min =  -266  Ave =    -4.49  Std =    19.37
   Min TM value:-266 at an offset of 127.838 usec -> 2021-07-20T09:08:43.723.186.562 (XO: 16.723189821)
   Max TM value:204 at an offset of 127.525 usec -> 2021-07-20T09:08:43.723.186.250 (XO: 16.723189509)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 130.213 usec -> 2021-07-20T09:08:43.723.188.937 (XO: 16.723192196)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 127.625 usec -> 2021-07-20T09:08:43.723.186.350 (XO: 16.723189609)

 - Block number: 3 starting at 2021-07-20T09:09:07.974.082.725 (XO: 40.974086637 - 40.974392638, t=37.2 degC)  Rank: 00085
   TM values: Max =   233  Min =  -264  Ave =    -4.51  Std =    19.59
   Min TM value:-264 at an offset of 125.688 usec -> 2021-07-20T09:09:07.974.208.412 (XO: 40.974212325)
   Max TM value:233 at an offset of 128.388 usec -> 2021-07-20T09:09:07.974.211.112 (XO: 40.974215025)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 123.463 usec -> 2021-07-20T09:09:07.974.206.187 (XO: 40.974210100)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 125.563 usec -> 2021-07-20T09:09:07.974.208.287 (XO: 40.974212200)

 - Block number: 4 starting at 2021-07-20T09:09:23.364.519.725 (XO: 56.364521928 - 56.364827929, t=37.2 degC)  Rank: 00081
   TM values: Max =   236  Min =  -245  Ave =    -4.46  Std =    25.38
   Min TM value:-245 at an offset of 130.188 usec -> 2021-07-20T09:09:23.364.649.912 (XO: 56.364652116)
   Max TM value:236 at an offset of 129.938 usec -> 2021-07-20T09:09:23.364.649.662 (XO: 56.364651866)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 132.575 usec -> 2021-07-20T09:09:23.364.652.300 (XO: 56.364654503)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 130.050 usec -> 2021-07-20T09:09:23.364.649.775 (XO: 56.364651978)

 - Block number: 5 starting at 2021-07-20T09:09:34.534.972.725 (XO: 67.534975244 - 67.535281245, t=37.2 degC)  Rank: 00083
   TM values: Max =   223  Min =  -273  Ave =    -4.56  Std =    21.01
   Min TM value:-273 at an offset of 40.638 usec -> 2021-07-20T09:09:34.535.013.362 (XO: 67.535015882)
   Max TM value:223 at an offset of 38.613 usec -> 2021-07-20T09:09:34.535.011.337 (XO: 67.535013857)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 38.438 usec -> 2021-07-20T09:09:34.535.011.162 (XO: 67.535013682)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 40.488 usec -> 2021-07-20T09:09:34.535.013.212 (XO: 67.535015732)
