-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov 18 05:16:14 2021 UTC
-- File 20210719_044912_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.832624 to 57.333161
-each separately from 2s to 57s and from 212739us to 832913us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.879683 in 1 packets: 18
      13.499854 in 1 packets: 36
      16.119844 in 1 packets: 54
MIU packet time from 2021-07-19T04:49:17.832.624.000 to 2021-07-19T04:50:12.333.161.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.879683 in 1 packets: 18
      13.499854 in 1 packets: 36
      16.119844 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8192362 to 58.8190518
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999161 in 1 packets: 18
      13.9999530 in 1 packets: 36
      15.9999465 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210719_044912_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6518610 to 4.6521500
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.8795990  -- OK for SL --
   MIU time from 2021-07-19T04:49:17.832.624.000 to 2021-07-19T04:49:17.832.913.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.879683

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5317490 to 29.5320380
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4998070  -- OK for SL --
   MIU time from 2021-07-19T04:49:42.712.596.000 to 2021-07-19T04:49:42.712.885.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.499854

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.0318450 to 43.0321340
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1197910  -- OK for SL --
   MIU time from 2021-07-19T04:49:56.212.739.000 to 2021-07-19T04:49:56.213.028.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.119844

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.1519250 to 59.1522140
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-19T04:50:12.332.872.000 to 2021-07-19T04:50:12.333.161.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.651861000 corrected by 40bit 10MHz counter of 3.819236200 obtained at 1Hz counter of 2. Result: 2.832627631
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-19T04:49:17.832.483.725
   Max 1st sample MIU time among all SL packets: 2021-07-19T04:50:12.333.020.725
   Min 1st sample XO time among all SL packets: 2.832487355
   Max 1st sample XO time among all SL packets: 57.333023057
   TM values from -272 to 251 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-19T04:49:17.832.483.725 (XO: 2.832487355 - 2.832793356, t=37.2 degC)  Rank: 00081
   TM values: Max =   223  Min =  -265  Ave =    -4.39  Std =    23.63
   Min TM value:-265 at an offset of 105.925 usec -> 2021-07-19T04:49:17.832.589.650 (XO: 2.832593281)
   Max TM value:223 at an offset of 105.625 usec -> 2021-07-19T04:49:17.832.589.350 (XO: 2.832592981)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 108.275 usec -> 2021-07-19T04:49:17.832.592.000 (XO: 2.832595631)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 105.725 usec -> 2021-07-19T04:49:17.832.589.450 (XO: 2.832593081)

 - Block number: 2 starting at 2021-07-19T04:49:42.712.455.725 (XO: 27.712458847 - 27.712764848, t=37.2 degC)  Rank: 00078
   TM values: Max =   213  Min =  -261  Ave =    -4.32  Std =    20.54
   Min TM value:-261 at an offset of 120.050 usec -> 2021-07-19T04:49:42.712.575.775 (XO: 27.712578898)
   Max TM value:213 at an offset of 122.600 usec -> 2021-07-19T04:49:42.712.578.325 (XO: 27.712581448)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 122.413 usec -> 2021-07-19T04:49:42.712.578.137 (XO: 27.712581260)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 119.863 usec -> 2021-07-19T04:49:42.712.575.587 (XO: 27.712578710)

 - Block number: 3 starting at 2021-07-19T04:49:56.212.598.725 (XO: 41.212600148 - 41.212906149, t=37.2 degC)  Rank: 00087
   TM values: Max =   231  Min =  -250  Ave =    -4.37  Std =    21.27
   Min TM value:-250 at an offset of 108.575 usec -> 2021-07-19T04:49:56.212.707.300 (XO: 41.212708723)
   Max TM value:231 at an offset of 105.988 usec -> 2021-07-19T04:49:56.212.704.712 (XO: 41.212706136)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 108.663 usec -> 2021-07-19T04:49:56.212.707.387 (XO: 41.212708811)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 106.088 usec -> 2021-07-19T04:49:56.212.704.812 (XO: 41.212706236)

 - Block number: 4 starting at 2021-07-19T04:50:12.332.731.725 (XO: 57.332734056 - 57.333040057, t=37.2 degC)  Rank: 00080
   TM values: Max =   251  Min =  -272  Ave =    -4.38  Std =    22.73
   Min TM value:-272 at an offset of 119.388 usec -> 2021-07-19T04:50:12.332.851.112 (XO: 57.332853444)
   Max TM value:251 at an offset of 116.850 usec -> 2021-07-19T04:50:12.332.848.575 (XO: 57.332850907)
   Max dTM/dt [1/us]:2.47e+03 at an offset of 119.488 usec -> 2021-07-19T04:50:12.332.851.212 (XO: 57.332853544)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 116.938 usec -> 2021-07-19T04:50:12.332.848.662 (XO: 57.332850994)
