-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov 18 04:56:21 2021 UTC
-- File 20210719_043741_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.827789 to 39.918325
-each separately from 13s to 39s and from 797938us to 918325us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.969860 in 1 packets: 18
      15.119809 in 1 packets: 36
MIU packet time from 2021-07-19T04:37:57.827.789.000 to 2021-07-19T04:38:23.918.325.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.969860 in 1 packets: 18
      15.119809 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8744310 to 40.8743439
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999632 in 1 packets: 18
      14.9999497 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210719_043741_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.7022208 to 15.7025098
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9698230  -- OK for SL --
   MIU time from 2021-07-19T04:37:57.827.789.000 to 2021-07-19T04:37:57.828.078.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.969860

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6723328 to 26.6726218
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1197590  -- OK for SL --
   MIU time from 2021-07-19T04:38:08.797.938.000 to 2021-07-19T04:38:08.798.227.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.119809

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.7923808 to 41.7926698
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-19T04:38:23.918.036.000 to 2021-07-19T04:38:23.918.325.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.702220800 corrected by 40bit 10MHz counter of 14.874431000 obtained at 1Hz counter of 13. Result: 13.827792532
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-19T04:37:57.827.648.725
   Max 1st sample MIU time among all SL packets: 2021-07-19T04:38:23.918.184.725
   Min 1st sample XO time among all SL packets: 13.827652256
   Max 1st sample XO time among all SL packets: 39.918188747
   TM values from -254 to 248 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-19T04:37:57.827.648.725 (XO: 13.827652256 - 13.827958257, t=37.2 degC)  Rank: 00077
   TM values: Max =   211  Min =  -238  Ave =    -4.42  Std =    26.10
   Min TM value:-238 at an offset of 113.988 usec -> 2021-07-19T04:37:57.827.762.712 (XO: 13.827766244)
   Max TM value:211 at an offset of 128.825 usec -> 2021-07-19T04:37:57.827.777.550 (XO: 13.827781082)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 128.638 usec -> 2021-07-19T04:37:57.827.777.362 (XO: 13.827780894)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 113.900 usec -> 2021-07-19T04:37:57.827.762.625 (XO: 13.827766157)

 - Block number: 2 starting at 2021-07-19T04:38:08.797.797.725 (XO: 24.797801038 - 24.798107039, t=37.2 degC)  Rank: 00091
   TM values: Max =   248  Min =  -254  Ave =    -4.51  Std =    20.50
   Min TM value:-254 at an offset of 126.088 usec -> 2021-07-19T04:38:08.797.923.812 (XO: 24.797927125)
   Max TM value:248 at an offset of 128.800 usec -> 2021-07-19T04:38:08.797.926.525 (XO: 24.797929838)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 128.688 usec -> 2021-07-19T04:38:08.797.926.412 (XO: 24.797929725)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 125.963 usec -> 2021-07-19T04:38:08.797.923.687 (XO: 24.797927000)

 - Block number: 3 starting at 2021-07-19T04:38:23.917.895.725 (XO: 39.917899746 - 39.918205747, t=37.2 degC)  Rank: 00083
   TM values: Max =   242  Min =  -234  Ave =    -4.40  Std =    19.75
   Min TM value:-234 at an offset of 126.063 usec -> 2021-07-19T04:38:23.918.021.787 (XO: 39.918025809)
   Max TM value:242 at an offset of 128.788 usec -> 2021-07-19T04:38:23.918.024.512 (XO: 39.918028534)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 128.688 usec -> 2021-07-19T04:38:23.918.024.412 (XO: 39.918028434)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 112.913 usec -> 2021-07-19T04:38:23.918.008.637 (XO: 39.918012659)
