-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Nov 18 01:56:01 2021 UTC
-- File 20210719_014719_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.059966 to 25.268380
-each separately from 2s to 25s and from 59966us to 268380us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.207836 in 1 packets: 18
MIU packet time from 2021-07-19T01:47:24.059.966.000 to 2021-07-19T01:47:47.268.380.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.207836 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8289178 to 26.8288409
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999231 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210719_014719_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.8888851 to 3.8891741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.2077590  -- OK for SL --
   MIU time from 2021-07-19T01:47:24.059.966.000 to 2021-07-19T01:47:24.060.255.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.207836

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0969331 to 27.0972221
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-19T01:47:47.268.091.000 to 2021-07-19T01:47:47.268.380.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.888885100 corrected by 40bit 10MHz counter of 3.828917800 obtained at 1Hz counter of 2. Result: 2.059967498
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-19T01:47:24.059.825.725
   Max 1st sample MIU time among all SL packets: 2021-07-19T01:47:47.268.239.725
   Min 1st sample XO time among all SL packets: 2.059827222
   Max 1st sample XO time among all SL packets: 25.268241810
   TM values from -259 to 263 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-19T01:47:24.059.825.725 (XO: 2.059827222 - 2.060133223, t=37.2 degC)  Rank: 00080
   TM values: Max =   263  Min =  -259  Ave =    -4.47  Std =    23.70
   Min TM value:-259 at an offset of 123.050 usec -> 2021-07-19T01:47:24.059.948.775 (XO: 2.059950273)
   Max TM value:263 at an offset of 125.763 usec -> 2021-07-19T01:47:24.059.951.487 (XO: 2.059952985)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 125.638 usec -> 2021-07-19T01:47:24.059.951.362 (XO: 2.059952860)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 122.938 usec -> 2021-07-19T01:47:24.059.948.662 (XO: 2.059950160)

 - Block number: 2 starting at 2021-07-19T01:47:47.267.950.725 (XO: 25.267952809 - 25.268258810, t=37.2 degC)  Rank: 00087
   TM values: Max =   228  Min =  -258  Ave =    -4.45  Std =    22.50
   Min TM value:-258 at an offset of 121.800 usec -> 2021-07-19T01:47:47.268.072.525 (XO: 25.268074610)
   Max TM value:228 at an offset of 124.438 usec -> 2021-07-19T01:47:47.268.075.162 (XO: 25.268077247)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 124.225 usec -> 2021-07-19T01:47:47.268.074.950 (XO: 25.268077035)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 121.638 usec -> 2021-07-19T01:47:47.268.072.362 (XO: 25.268074447)
