-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 16 18:59:03 2021 UTC
-- File 20210717_210419_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.249846 to 22.070426
-each separately from 7s to 22s and from 70137us to 250135us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.820002 in 1 packets: 18
MIU packet time from 2021-07-17T21:04:29.249.846.000 to 2021-07-17T21:04:44.070.426.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.820002 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9046307 to 23.9045803
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999496 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210717_210419_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.1544781 to 9.1547671
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8199510  -- OK for SL --
   MIU time from 2021-07-17T21:04:29.249.846.000 to 2021-07-17T21:04:29.250.135.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.820002

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.9747181 to 23.9750071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-17T21:04:44.070.137.000 to 2021-07-17T21:04:44.070.426.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.154478100 corrected by 40bit 10MHz counter of 8.904630700 obtained at 1Hz counter of 7. Result: 7.249848249
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-17T21:04:29.249.705.725
   Max 1st sample MIU time among all SL packets: 2021-07-17T21:04:44.070.285.725
   Min 1st sample XO time among all SL packets: 7.249707974
   Max 1st sample XO time among all SL packets: 22.070286764
   TM values from -287 to 243 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-17T21:04:29.249.705.725 (XO: 7.249707974 - 7.250013975, t=37.2 degC)  Rank: 00085
   TM values: Max =   221  Min =  -285  Ave =    -4.47  Std =    27.20
   Min TM value:-285 at an offset of 116.500 usec -> 2021-07-17T21:04:29.249.822.225 (XO: 7.249824474)
   Max TM value:221 at an offset of 119.038 usec -> 2021-07-17T21:04:29.249.824.762 (XO: 7.249827012)
   Max dTM/dt [1/us]:2.73e+03 at an offset of 118.863 usec -> 2021-07-17T21:04:29.249.824.587 (XO: 7.249826837)
   Min dTM/dt [1/us]:-3.08e+03 at an offset of 116.325 usec -> 2021-07-17T21:04:29.249.822.050 (XO: 7.249824299)

 - Block number: 2 starting at 2021-07-17T21:04:44.069.996.725 (XO: 22.069997763 - 22.070303764, t=37.2 degC)  Rank: 00088
   TM values: Max =   243  Min =  -287  Ave =    -4.53  Std =    24.61
   Min TM value:-287 at an offset of 103.900 usec -> 2021-07-17T21:04:44.070.100.625 (XO: 22.070101663)
   Max TM value:243 at an offset of 119.125 usec -> 2021-07-17T21:04:44.070.115.850 (XO: 22.070116888)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 118.925 usec -> 2021-07-17T21:04:44.070.115.650 (XO: 22.070116688)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 116.350 usec -> 2021-07-17T21:04:44.070.113.075 (XO: 22.070114113)
