-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov 16 18:10:55 2021 UTC
-- File 20210717_202411_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.957963 to 24.474510
-each separately from 9s to 24s and from 474221us to 958252us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.515969 in 1 packets: 18
MIU packet time from 2021-07-17T20:24:23.957.963.000 to 2021-07-17T20:24:38.474.510.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.515969 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8991177 to 25.8990674
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999497 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210717_202411_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8570816 to 11.8573706
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5159190  -- OK for SL --
   MIU time from 2021-07-17T20:24:23.957.963.000 to 2021-07-17T20:24:23.958.252.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.515969

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3732896 to 26.3735786
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-17T20:24:38.474.221.000 to 2021-07-17T20:24:38.474.510.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.857081600 corrected by 40bit 10MHz counter of 10.899117700 obtained at 1Hz counter of 9. Result: 9.957967061
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-17T20:24:23.957.822.725
   Max 1st sample MIU time among all SL packets: 2021-07-17T20:24:38.474.369.725
   Min 1st sample XO time among all SL packets: 9.957826786
   Max 1st sample XO time among all SL packets: 24.474372490
   TM values from -286 to 255 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-17T20:24:23.957.822.725 (XO: 9.957826786 - 9.958132787, t=37.2 degC)  Rank: 00081
   TM values: Max =   255  Min =  -286  Ave =    -4.55  Std =    26.78
   Min TM value:-286 at an offset of 122.550 usec -> 2021-07-17T20:24:23.957.945.275 (XO: 9.957949336)
   Max TM value:255 at an offset of 122.300 usec -> 2021-07-17T20:24:23.957.945.025 (XO: 9.957949086)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 120.300 usec -> 2021-07-17T20:24:23.957.943.025 (XO: 9.957947086)
   Min dTM/dt [1/us]:-2.68e+03 at an offset of 122.425 usec -> 2021-07-17T20:24:23.957.945.150 (XO: 9.957949211)

 - Block number: 2 starting at 2021-07-17T20:24:38.474.080.725 (XO: 24.474083489 - 24.474389490, t=37.2 degC)  Rank: 00083
   TM values: Max =   244  Min =  -253  Ave =    -4.42  Std =    21.59
   Min TM value:-253 at an offset of 118.488 usec -> 2021-07-17T20:24:38.474.199.212 (XO: 24.474201977)
   Max TM value:244 at an offset of 121.200 usec -> 2021-07-17T20:24:38.474.201.925 (XO: 24.474204690)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 121.125 usec -> 2021-07-17T20:24:38.474.201.850 (XO: 24.474204615)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 118.363 usec -> 2021-07-17T20:24:38.474.199.087 (XO: 24.474201852)
