-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Nov 14 21:06:30 2021 UTC
-- File 20210714_203537_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.368992 to 29.064092
-each separately from 2s to 29s and from 63803us to 369281us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.694522 in 1 packets: 18
MIU packet time from 2021-07-14T20:35:42.368.992.000 to 2021-07-14T20:36:09.064.092.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.694522 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8297664 to 30.8296754
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999090 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210714_203537_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1987595 to 4.1990485
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.6944310  -- OK for SL --
   MIU time from 2021-07-14T20:35:42.368.992.000 to 2021-07-14T20:35:42.369.281.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.694522

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      95 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.8934795 to 30.8937685
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-14T20:36:09.063.803.000 to 2021-07-14T20:36:09.064.092.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.198759500 corrected by 40bit 10MHz counter of 3.829766400 obtained at 1Hz counter of 2. Result: 2.368994355
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-14T20:35:42.368.851.725
   Max 1st sample MIU time among all SL packets: 2021-07-14T20:36:09.063.951.725
   Min 1st sample XO time among all SL packets: 2.368854079
   Max 1st sample XO time among all SL packets: 29.063953042
   TM values from -299 to 270 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-14T20:35:42.368.851.725 (XO: 2.368854079 - 2.369160080, t=37.2 degC)  Rank: 00078
   TM values: Max =   227  Min =  -278  Ave =    -4.47  Std =    24.30
   Min TM value:-278 at an offset of 125.575 usec -> 2021-07-14T20:35:42.368.977.300 (XO: 2.368979655)
   Max TM value:227 at an offset of 128.138 usec -> 2021-07-14T20:35:42.368.979.862 (XO: 2.368982217)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 127.975 usec -> 2021-07-14T20:35:42.368.979.700 (XO: 2.368982055)
   Min dTM/dt [1/us]:-2.07e+03 at an offset of 125.425 usec -> 2021-07-14T20:35:42.368.977.150 (XO: 2.368979505)

 - Block number: 2 starting at 2021-07-14T20:36:09.063.662.725 (XO: 29.063664041 - 29.063970042, t=37.2 degC)  Rank: 00095
   TM values: Max =   270  Min =  -299  Ave =    -4.60  Std =    26.49
   Min TM value:-299 at an offset of 123.138 usec -> 2021-07-14T20:36:09.063.785.862 (XO: 29.063787179)
   Max TM value:270 at an offset of 110.425 usec -> 2021-07-14T20:36:09.063.773.150 (XO: 29.063774467)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 108.400 usec -> 2021-07-14T20:36:09.063.771.125 (XO: 29.063772442)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 110.550 usec -> 2021-07-14T20:36:09.063.773.275 (XO: 29.063774592)
