-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Nov 14 18:24:16 2021 UTC
-- File 20210714_184841_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.542905 to 62.330888
-each separately from 1s to 62s and from 121212us to 745310us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.804407 in 1 packets: 18
      13.397131 in 1 packets: 36
      18.375902 in 1 packets: 54
      3.209098 in 1 packets: 72
MIU packet time from 2021-07-14T18:48:45.542.905.000 to 2021-07-14T18:49:46.330.888.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.804407 in 1 packets: 18
      13.397131 in 1 packets: 36
      18.375902 in 1 packets: 54
      3.209098 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8894924 to 63.8892871
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999125 in 1 packets: 18
      12.9999562 in 1 packets: 36
      18.9999361 in 1 packets: 54
      2.9999899 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210714_184841_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4323986 to 3.4326876
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.8043190  -- OK for SL --
   MIU time from 2021-07-14T18:48:45.542.905.000 to 2021-07-14T18:48:45.543.194.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.804407

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.2370066 to 29.2372956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3970870  -- OK for SL --
   MIU time from 2021-07-14T18:49:11.347.601.000 to 2021-07-14T18:49:11.347.890.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.397131

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.6343826 to 42.6346716
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3758390  -- OK for SL --
   MIU time from 2021-07-14T18:49:24.745.021.000 to 2021-07-14T18:49:24.745.310.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.375902

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.0105106 to 61.0107996
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.2090870  -- OK for SL --
   MIU time from 2021-07-14T18:49:43.121.212.000 to 2021-07-14T18:49:43.121.501.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.209098

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 64.2198866 to 64.2201756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-14T18:49:46.330.599.000 to 2021-07-14T18:49:46.330.888.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.432398600 corrected by 40bit 10MHz counter of 2.889492400 obtained at 1Hz counter of 1. Result: 1.542907992
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-14T18:48:45.542.764.725
   Max 1st sample MIU time among all SL packets: 2021-07-14T18:49:46.330.747.725
   Min 1st sample XO time among all SL packets: 1.542767716
   Max 1st sample XO time among all SL packets: 62.330749350
   TM values from -290 to 278 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-14T18:48:45.542.764.725 (XO: 1.542767716 - 1.543073717, t=37.2 degC)  Rank: 00084
   TM values: Max =   217  Min =  -261  Ave =    -4.55  Std =    23.64
   Min TM value:-261 at an offset of 103.138 usec -> 2021-07-14T18:48:45.542.867.862 (XO: 1.542870854)
   Max TM value:217 at an offset of 118.463 usec -> 2021-07-14T18:48:45.542.883.187 (XO: 1.542886179)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 118.363 usec -> 2021-07-14T18:48:45.542.883.087 (XO: 1.542886079)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 102.925 usec -> 2021-07-14T18:48:45.542.867.650 (XO: 1.542870641)

 - Block number: 2 starting at 2021-07-14T18:49:11.347.460.725 (XO: 27.347462606 - 27.347768607, t=37.2 degC)  Rank: 00079
   TM values: Max =   278  Min =  -278  Ave =    -4.54  Std =    32.66
   Min TM value:-278 at an offset of 118.763 usec -> 2021-07-14T18:49:11.347.579.487 (XO: 27.347581369)
   Max TM value:278 at an offset of 160.801 usec -> 2021-07-14T18:49:11.347.621.525 (XO: 27.347623407)
   Max dTM/dt [1/us]:2.63e+03 at an offset of 160.663 usec -> 2021-07-14T18:49:11.347.621.388 (XO: 27.347623269)
   Min dTM/dt [1/us]:-2.59e+03 at an offset of 162.801 usec -> 2021-07-14T18:49:11.347.623.525 (XO: 27.347625407)

 - Block number: 3 starting at 2021-07-14T18:49:24.744.880.725 (XO: 40.744883683 - 40.745189684, t=37.2 degC)  Rank: 00080
   TM values: Max =   244  Min =  -257  Ave =    -4.55  Std =    33.73
   Min TM value:-257 at an offset of 129.638 usec -> 2021-07-14T18:49:24.745.010.362 (XO: 40.745013321)
   Max TM value:244 at an offset of 127.625 usec -> 2021-07-14T18:49:24.745.008.350 (XO: 40.745011309)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 114.600 usec -> 2021-07-14T18:49:24.744.995.325 (XO: 40.744998283)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 116.738 usec -> 2021-07-14T18:49:24.744.997.462 (XO: 40.745000421)

 - Block number: 4 starting at 2021-07-14T18:49:43.121.071.725 (XO: 59.121073537 - 59.121379538, t=37.2 degC)  Rank: 00089
   TM values: Max =   226  Min =  -255  Ave =    -4.41  Std =    23.30
   Min TM value:-255 at an offset of 128.200 usec -> 2021-07-14T18:49:43.121.199.925 (XO: 59.121201737)
   Max TM value:226 at an offset of 130.925 usec -> 2021-07-14T18:49:43.121.202.650 (XO: 59.121204462)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 130.750 usec -> 2021-07-14T18:49:43.121.202.475 (XO: 59.121204287)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 128.038 usec -> 2021-07-14T18:49:43.121.199.762 (XO: 59.121201575)

 - Block number: 5 starting at 2021-07-14T18:49:46.330.458.725 (XO: 62.330460349 - 62.330766350, t=37.2 degC)  Rank: 00075
   TM values: Max =   208  Min =  -290  Ave =    -4.44  Std =    22.91
   Min TM value:-290 at an offset of 131.850 usec -> 2021-07-14T18:49:46.330.590.575 (XO: 62.330592199)
   Max TM value:208 at an offset of 131.538 usec -> 2021-07-14T18:49:46.330.590.262 (XO: 62.330591887)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 134.225 usec -> 2021-07-14T18:49:46.330.592.950 (XO: 62.330594574)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 131.638 usec -> 2021-07-14T18:49:46.330.590.362 (XO: 62.330591987)
