-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Nov 14 05:54:28 2021 UTC
-- File 20210714_085649_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.798594 to 28.446909
-each separately from 6s to 28s and from 446620us to 798883us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.647737 in 1 packets: 18
MIU packet time from 2021-07-14T08:56:58.798.594.000 to 2021-07-14T08:57:20.446.909.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.647737 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9009027 to 29.9008283
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999256 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210714_085649_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6994974 to 8.6997864
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6476630  -- OK for SL --
   MIU time from 2021-07-14T08:56:58.798.594.000 to 2021-07-14T08:56:58.798.883.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.647737

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.3474494 to 30.3477384
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-14T08:57:20.446.620.000 to 2021-07-14T08:57:20.446.909.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.699497400 corrected by 40bit 10MHz counter of 7.900902700 obtained at 1Hz counter of 6. Result: 6.798597335
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-14T08:56:58.798.453.725
   Max 1st sample MIU time among all SL packets: 2021-07-14T08:57:20.446.768.725
   Min 1st sample XO time among all SL packets: 6.798457060
   Max 1st sample XO time among all SL packets: 28.446771344
   TM values from -304 to 220 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-14T08:56:58.798.453.725 (XO: 6.798457060 - 6.798763061, t=37.2 degC)  Rank: 00076
   TM values: Max =   213  Min =  -258  Ave =    -4.46  Std =    22.33
   Min TM value:-258 at an offset of 120.263 usec -> 2021-07-14T08:56:58.798.573.987 (XO: 6.798577323)
   Max TM value:213 at an offset of 119.963 usec -> 2021-07-14T08:56:58.798.573.687 (XO: 6.798577023)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 122.663 usec -> 2021-07-14T08:56:58.798.576.387 (XO: 6.798579723)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 120.075 usec -> 2021-07-14T08:56:58.798.573.800 (XO: 6.798577135)

 - Block number: 2 starting at 2021-07-14T08:57:20.446.479.725 (XO: 28.446482343 - 28.446788344, t=37.2 degC)  Rank: 00084
   TM values: Max =   220  Min =  -304  Ave =    -4.47  Std =    27.27
   Min TM value:-304 at an offset of 106.200 usec -> 2021-07-14T08:57:20.446.585.925 (XO: 28.446588543)
   Max TM value:220 at an offset of 76.275 usec -> 2021-07-14T08:57:20.446.556.000 (XO: 28.446558618)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 108.563 usec -> 2021-07-14T08:57:20.446.588.287 (XO: 28.446590906)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 106.025 usec -> 2021-07-14T08:57:20.446.585.750 (XO: 28.446588368)
