-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 17:25:53 2021 UTC
-- File 20210712_230853_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.449266 to 24.159154
-each separately from 5s to 24s and from 158865us to 449555us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.709310 in 1 packets: 18
MIU packet time from 2021-07-12T23:09:01.449.266.000 to 2021-07-12T23:09:20.159.154.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.709310 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8937941 to 25.8937303
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999362 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_230853_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3430607 to 7.3433497
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7092470  -- OK for SL --
   MIU time from 2021-07-12T23:09:01.449.266.000 to 2021-07-12T23:09:01.449.555.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.709310

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.0525967 to 26.0528857
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T23:09:20.158.865.000 to 2021-07-12T23:09:20.159.154.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.343060700 corrected by 40bit 10MHz counter of 6.893794100 obtained at 1Hz counter of 5. Result: 5.449268128
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-12T23:09:01.449.125.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T23:09:20.159.013.725
   Min 1st sample XO time among all SL packets: 5.449127852
   Max 1st sample XO time among all SL packets: 24.159015650
   TM values from -281 to 226 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T23:09:01.449.125.725 (XO: 5.449127852 - 5.449433853, t=37.2 degC)  Rank: 00074
   TM values: Max =   210  Min =  -281  Ave =    -4.44  Std =    19.02
   Min TM value:-281 at an offset of 104.588 usec -> 2021-07-12T23:09:01.449.230.312 (XO: 5.449232440)
   Max TM value:210 at an offset of 104.300 usec -> 2021-07-12T23:09:01.449.230.025 (XO: 5.449232152)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 102.150 usec -> 2021-07-12T23:09:01.449.227.875 (XO: 5.449230002)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 104.400 usec -> 2021-07-12T23:09:01.449.230.125 (XO: 5.449232252)

 - Block number: 2 starting at 2021-07-12T23:09:20.158.724.725 (XO: 24.158726649 - 24.159032650, t=37.2 degC)  Rank: 00082
   TM values: Max =   226  Min =  -260  Ave =    -4.42  Std =    23.80
   Min TM value:-260 at an offset of 128.225 usec -> 2021-07-12T23:09:20.158.852.950 (XO: 24.158854874)
   Max TM value:226 at an offset of 127.938 usec -> 2021-07-12T23:09:20.158.852.662 (XO: 24.158854587)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 130.625 usec -> 2021-07-12T23:09:20.158.855.350 (XO: 24.158857274)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 128.050 usec -> 2021-07-12T23:09:20.158.852.775 (XO: 24.158854699)
