-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 12:46:49 2021 UTC
-- File 20210712_134948_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.225284 to 27.555785
-each separately from 2s to 27s and from 225284us to 555785us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.329923 in 1 packets: 18
MIU packet time from 2021-07-12T13:49:53.225.284.000 to 2021-07-12T13:50:18.555.785.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.329923 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4177922 to 28.4177084
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999162 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_134948_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.6430774 to 3.6433664
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.3298390  -- OK for SL --
   MIU time from 2021-07-12T13:49:53.225.284.000 to 2021-07-12T13:49:53.225.573.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.329923

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.9732054 to 28.9734944
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T13:50:18.555.496.000 to 2021-07-12T13:50:18.555.785.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.643077400 corrected by 40bit 10MHz counter of 3.417792200 obtained at 1Hz counter of 2. Result: 2.225285966
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-12T13:49:53.225.143.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T13:50:18.555.644.725
   Min 1st sample XO time among all SL packets: 2.225145690
   Max 1st sample XO time among all SL packets: 27.555647559
   TM values from -243 to 232 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T13:49:53.225.143.725 (XO: 2.225145690 - 2.225451692, t=37.2 degC)  Rank: 00080
   TM values: Max =   232  Min =  -239  Ave =    -4.61  Std =    18.72
   Min TM value:-239 at an offset of 121.825 usec -> 2021-07-12T13:49:53.225.265.550 (XO: 2.225267516)
   Max TM value:232 at an offset of 124.550 usec -> 2021-07-12T13:49:53.225.268.275 (XO: 2.225270241)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 124.413 usec -> 2021-07-12T13:49:53.225.268.137 (XO: 2.225270103)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 109.613 usec -> 2021-07-12T13:49:53.225.253.337 (XO: 2.225255303)

 - Block number: 2 starting at 2021-07-12T13:50:18.555.355.725 (XO: 27.555358558 - 27.555664559, t=37.2 degC)  Rank: 00081
   TM values: Max =   231  Min =  -243  Ave =    -4.48  Std =    27.48
   Min TM value:-243 at an offset of 156.088 usec -> 2021-07-12T13:50:18.555.511.813 (XO: 27.555514646)
   Max TM value:231 at an offset of 114.938 usec -> 2021-07-12T13:50:18.555.470.662 (XO: 27.555473496)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 179.838 usec -> 2021-07-12T13:50:18.555.535.563 (XO: 27.555538396)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 155.926 usec -> 2021-07-12T13:50:18.555.511.650 (XO: 27.555514483)
