-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 08:34:10 2021 UTC
-- File 20210712_095338_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.722517 to 44.602893
-each separately from 13s to 44s and from 52431us to 722806us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.329625 in 1 packets: 18
      23.549884 in 1 packets: 36
MIU packet time from 2021-07-12T09:53:54.722.517.000 to 2021-07-12T09:54:25.602.893.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.329625 in 1 packets: 18
      23.549884 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9233214 to 45.9232176
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999731 in 1 packets: 18
      22.9999231 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_095338_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.6458387 to 15.6461277
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.3295990  -- OK for SL --
   MIU time from 2021-07-12T09:53:54.722.517.000 to 2021-07-12T09:53:54.722.806.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.329625

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9757267 to 22.9760157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5498070  -- OK for SL --
   MIU time from 2021-07-12T09:54:02.052.431.000 to 2021-07-12T09:54:02.052.720.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.549884

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.5258227 to 46.5261117
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T09:54:25.602.604.000 to 2021-07-12T09:54:25.602.893.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.645838700 corrected by 40bit 10MHz counter of 14.923321400 obtained at 1Hz counter of 13. Result: 13.722519757
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-12T09:53:54.722.376.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T09:54:25.602.752.725
   Min 1st sample XO time among all SL packets: 13.722379481
   Max 1st sample XO time among all SL packets: 44.602755874
   TM values from -244 to 238 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T09:53:54.722.376.725 (XO: 13.722379481 - 13.722685482, t=37.2 degC)  Rank: 00084
   TM values: Max =   229  Min =  -234  Ave =    -4.54  Std =    20.33
   Min TM value:-234 at an offset of 108.713 usec -> 2021-07-12T09:53:54.722.485.437 (XO: 13.722488194)
   Max TM value:229 at an offset of 111.350 usec -> 2021-07-12T09:53:54.722.488.075 (XO: 13.722490831)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 111.200 usec -> 2021-07-12T09:53:54.722.487.925 (XO: 13.722490681)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 108.500 usec -> 2021-07-12T09:53:54.722.485.225 (XO: 13.722487981)

 - Block number: 2 starting at 2021-07-12T09:54:02.052.290.725 (XO: 21.052292098 - 21.052598099, t=37.2 degC)  Rank: 00079
   TM values: Max =   238  Min =  -244  Ave =    -4.58  Std =    17.08
   Min TM value:-244 at an offset of 109.200 usec -> 2021-07-12T09:54:02.052.399.925 (XO: 21.052401298)
   Max TM value:238 at an offset of 108.900 usec -> 2021-07-12T09:54:02.052.399.625 (XO: 21.052400998)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 111.588 usec -> 2021-07-12T09:54:02.052.402.312 (XO: 21.052403685)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 109.013 usec -> 2021-07-12T09:54:02.052.399.737 (XO: 21.052401110)

 - Block number: 3 starting at 2021-07-12T09:54:25.602.463.725 (XO: 44.602466873 - 44.602772874, t=37.2 degC)  Rank: 00082
   TM values: Max =   222  Min =  -222  Ave =    -4.47  Std =    21.26
   Min TM value:-222 at an offset of 109.063 usec -> 2021-07-12T09:54:25.602.572.787 (XO: 44.602575936)
   Max TM value:222 at an offset of 111.763 usec -> 2021-07-12T09:54:25.602.575.487 (XO: 44.602578636)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 111.625 usec -> 2021-07-12T09:54:25.602.575.350 (XO: 44.602578499)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 108.913 usec -> 2021-07-12T09:54:25.602.572.637 (XO: 44.602575786)
