-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 06:46:41 2021 UTC
-- File 20210712_090401_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.319299 to 19.714332
-each separately from 7s to 19s and from 319299us to 714332us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.394455 in 1 packets: 18
MIU packet time from 2021-07-12T09:04:11.319.299.000 to 2021-07-12T09:04:23.714.332.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.394455 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9350542 to 20.9350142
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999600 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_090401_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.2543540 to 9.2546430
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3944150  -- OK for SL --
   MIU time from 2021-07-12T09:04:11.319.299.000 to 2021-07-12T09:04:11.319.588.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.394455

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6490580 to 21.6493470
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T09:04:23.714.043.000 to 2021-07-12T09:04:23.714.332.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.254354000 corrected by 40bit 10MHz counter of 8.935054200 obtained at 1Hz counter of 7. Result: 7.319300854
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-12T09:04:11.319.158.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T09:04:23.714.191.725
   Min 1st sample XO time among all SL packets: 7.319160578
   Max 1st sample XO time among all SL packets: 19.714194882
   TM values from -240 to 214 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T09:04:11.319.158.725 (XO: 7.319160578 - 7.319466579, t=37.2 degC)  Rank: 00084
   TM values: Max =   214  Min =  -240  Ave =    -4.36  Std =    19.35
   Min TM value:-240 at an offset of 122.738 usec -> 2021-07-12T09:04:11.319.281.462 (XO: 7.319283316)
   Max TM value:214 at an offset of 120.163 usec -> 2021-07-12T09:04:11.319.278.887 (XO: 7.319280741)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 122.850 usec -> 2021-07-12T09:04:11.319.281.575 (XO: 7.319283429)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 120.275 usec -> 2021-07-12T09:04:11.319.279.000 (XO: 7.319280854)

 - Block number: 2 starting at 2021-07-12T09:04:23.713.902.725 (XO: 19.713905881 - 19.714211882, t=37.2 degC)  Rank: 00083
   TM values: Max =   214  Min =  -225  Ave =    -4.46  Std =    20.44
   Min TM value:-225 at an offset of 116.388 usec -> 2021-07-12T09:04:23.714.019.112 (XO: 19.714022269)
   Max TM value:214 at an offset of 119.050 usec -> 2021-07-12T09:04:23.714.021.775 (XO: 19.714024931)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 118.913 usec -> 2021-07-12T09:04:23.714.021.637 (XO: 19.714024794)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 116.200 usec -> 2021-07-12T09:04:23.714.018.925 (XO: 19.714022081)
