-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 06:05:09 2021 UTC
-- File 20210712_075558_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.176716 to 51.904727
-each separately from 1s to 51s and from 0us to 999980us.
   Differences in neigboring packets: 
      0.000017 in 67 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.822907 in 1 packets: 18
      0.000020 in 1 packets: 23
      11.037108 in 1 packets: 36
      11.866837 in 1 packets: 54
MIU packet time from 2021-07-12T07:56:02.176.716.000 to 2021-07-12T07:56:52.904.727.000
   Differences in neigboring packets: 
      0.000017 in 67 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.822907 in 1 packets: 18
      0.000020 in 1 packets: 23
      11.037108 in 1 packets: 36
      11.866837 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7096648 to 52.7094962
   Differences in neigboring packets: 
      0.0000000 in 67 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999089 in 1 packets: 18
      0.9999966 in 1 packets: 23
      10.9999629 in 1 packets: 36
      10.9999630 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 48 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 24 packets: 19, 20, 21, 22, 23, 24, 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_075558_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8863823 to 2.8866713
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.8228150  -- OK for SL --
   MIU time from 2021-07-12T07:56:02.176.716.000 to 2021-07-12T07:56:02.177.005.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.822907

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.7094863 to 30.7097753
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0370710  -- OK for SL --
   MIU time from 2021-07-12T07:56:29.999.912.000 to 2021-07-12T07:56:30.000.204.000
      Difference between neigboring packets: 
      0.000017 in 16 packets: 19, 20, 21, 22, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35
      0.000020 in 1 packets: 23
      Difference to the following block: 11.037108

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.7468463 to 41.7471353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8667990  -- OK for SL --
   MIU time from 2021-07-12T07:56:41.037.312.000 to 2021-07-12T07:56:41.037.601.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.866837

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.6139343 to 53.6142233
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T07:56:52.904.438.000 to 2021-07-12T07:56:52.904.727.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.886382300 corrected by 40bit 10MHz counter of 2.709664800 obtained at 1Hz counter of 1. Result: 1.176718101
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-12T07:56:02.176.575.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T07:56:52.904.586.725
   Min 1st sample XO time among all SL packets: 1.176577825
   Max 1st sample XO time among all SL packets: 51.904589901
   TM values from -298 to 263 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T07:56:02.176.575.725 (XO: 1.176577825 - 1.176883826, t=37.2 degC)  Rank: 00076
   TM values: Max =   248  Min =  -298  Ave =    -4.51  Std =    31.02
   Min TM value:-298 at an offset of 215.638 usec -> 2021-07-12T07:56:02.176.791.363 (XO: 1.176793464)
   Max TM value:248 at an offset of 124.325 usec -> 2021-07-12T07:56:02.176.700.050 (XO: 1.176702151)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 124.213 usec -> 2021-07-12T07:56:02.176.699.937 (XO: 1.176702038)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 215.513 usec -> 2021-07-12T07:56:02.176.791.238 (XO: 1.176793339)

 - Block number: 2 starting at 2021-07-12T07:56:29.999.771.725 (XO: 28.999775724 - 29.000081725, t=37.2 degC)  Rank: 00089
   TM values: Max =   257  Min =  -276  Ave =    -4.46  Std =    25.39
   Min TM value:-276 at an offset of 124.563 usec -> 2021-07-12T07:56:29.999.896.287 (XO: 28.999900287)
   Max TM value:257 at an offset of 121.975 usec -> 2021-07-12T07:56:29.999.893.700 (XO: 28.999897700)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 124.638 usec -> 2021-07-12T07:56:29.999.896.362 (XO: 28.999900362)
   Min dTM/dt [1/us]:-2.82e+03 at an offset of 122.100 usec -> 2021-07-12T07:56:29.999.893.825 (XO: 28.999897825)

 - Block number: 3 starting at 2021-07-12T07:56:41.037.171.725 (XO: 40.037172948 - 40.037478949, t=37.2 degC)  Rank: 00082
   TM values: Max =   226  Min =  -250  Ave =    -4.54  Std =    21.99
   Min TM value:-250 at an offset of 109.200 usec -> 2021-07-12T07:56:41.037.280.925 (XO: 40.037282148)
   Max TM value:226 at an offset of 111.850 usec -> 2021-07-12T07:56:41.037.283.575 (XO: 40.037284798)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 111.663 usec -> 2021-07-12T07:56:41.037.283.387 (XO: 40.037284611)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 108.988 usec -> 2021-07-12T07:56:41.037.280.712 (XO: 40.037281936)

 - Block number: 4 starting at 2021-07-12T07:56:52.904.297.725 (XO: 51.904300900 - 51.904606901, t=37.2 degC)  Rank: 00078
   TM values: Max =   263  Min =  -242  Ave =    -4.43  Std =    22.25
   Min TM value:-242 at an offset of 96.888 usec -> 2021-07-12T07:56:52.904.394.612 (XO: 51.904397787)
   Max TM value:263 at an offset of 99.625 usec -> 2021-07-12T07:56:52.904.397.350 (XO: 51.904400525)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 99.500 usec -> 2021-07-12T07:56:52.904.397.225 (XO: 51.904400400)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 96.738 usec -> 2021-07-12T07:56:52.904.394.462 (XO: 51.904397637)
