-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 04:16:37 2021 UTC
-- File 20210712_050628_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.555890 to 15.356352
-each separately from 11s to 15s and from 356063us to 556179us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.799884 in 1 packets: 18
MIU packet time from 2021-07-12T05:06:42.555.890.000 to 2021-07-12T05:06:46.356.352.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.799884 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8052333 to 16.8052198
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999865 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_050628_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3611238 to 13.3614128
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.7998710  -- OK for SL --
   MIU time from 2021-07-12T05:06:42.555.890.000 to 2021-07-12T05:06:42.556.179.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.799884

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1612838 to 17.1615728
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T05:06:46.356.063.000 to 2021-07-12T05:06:46.356.352.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.361123800 corrected by 40bit 10MHz counter of 12.805233300 obtained at 1Hz counter of 11. Result: 11.555892390
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-12T05:06:42.555.749.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T05:06:46.356.211.725
   Min 1st sample XO time among all SL packets: 11.555752115
   Max 1st sample XO time among all SL packets: 15.356213901
   TM values from -253 to 211 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T05:06:42.555.749.725 (XO: 11.555752115 - 11.556058116, t=37.2 degC)  Rank: 00081
   TM values: Max =   210  Min =  -221  Ave =    -4.47  Std =    18.45
   Min TM value:-221 at an offset of 119.425 usec -> 2021-07-12T05:06:42.555.869.150 (XO: 11.555871540)
   Max TM value:210 at an offset of 122.150 usec -> 2021-07-12T05:06:42.555.871.875 (XO: 11.555874265)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 122.025 usec -> 2021-07-12T05:06:42.555.871.750 (XO: 11.555874140)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 119.213 usec -> 2021-07-12T05:06:42.555.868.937 (XO: 11.555871327)

 - Block number: 2 starting at 2021-07-12T05:06:46.355.922.725 (XO: 15.355924900 - 15.356230901, t=37.2 degC)  Rank: 00087
   TM values: Max =   211  Min =  -253  Ave =    -4.49  Std =    24.94
   Min TM value:-253 at an offset of 119.725 usec -> 2021-07-12T05:06:46.356.042.450 (XO: 15.356044625)
   Max TM value:211 at an offset of 122.363 usec -> 2021-07-12T05:06:46.356.045.087 (XO: 15.356047262)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 122.200 usec -> 2021-07-12T05:06:46.356.044.925 (XO: 15.356047100)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 119.525 usec -> 2021-07-12T05:06:46.356.042.250 (XO: 15.356044425)
