-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Nov 12 02:48:46 2021 UTC
-- File 20210712_025943_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.365263 to 32.785876
-each separately from 12s to 32s and from 365263us to 785876us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.239898 in 1 packets: 18
      12.179848 in 1 packets: 36
MIU packet time from 2021-07-12T02:59:58.365.263.000 to 2021-07-12T03:00:18.785.876.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.239898 in 1 packets: 18
      12.179848 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9339856 to 33.9339183
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999731 in 1 packets: 18
      11.9999596 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210712_025943_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.2992495 to 14.2995385
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2398710  -- OK for SL --
   MIU time from 2021-07-12T02:59:58.365.263.000 to 2021-07-12T02:59:58.365.552.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.239898

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.5394095 to 22.5396985
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1798070  -- OK for SL --
   MIU time from 2021-07-12T03:00:06.605.450.000 to 2021-07-12T03:00:06.605.739.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.179848

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.7195055 to 34.7197945
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-12T03:00:18.785.587.000 to 2021-07-12T03:00:18.785.876.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.299249500 corrected by 40bit 10MHz counter of 13.933985600 obtained at 1Hz counter of 12. Result: 12.365265142
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-12T02:59:58.365.122.725
   Max 1st sample MIU time among all SL packets: 2021-07-12T03:00:18.785.735.725
   Min 1st sample XO time among all SL packets: 12.365124866
   Max 1st sample XO time among all SL packets: 32.785738597
   TM values from -288 to 273 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-12T02:59:58.365.122.725 (XO: 12.365124866 - 12.365430867, t=37.2 degC)  Rank: 00080
   TM values: Max =   273  Min =  -288  Ave =    -4.58  Std =    31.99
   Min TM value:-288 at an offset of 147.113 usec -> 2021-07-12T02:59:58.365.269.838 (XO: 12.365271979)
   Max TM value:273 at an offset of 119.538 usec -> 2021-07-12T02:59:58.365.242.262 (XO: 12.365244404)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 117.500 usec -> 2021-07-12T02:59:58.365.240.225 (XO: 12.365242367)
   Min dTM/dt [1/us]:-2.73e+03 at an offset of 119.650 usec -> 2021-07-12T02:59:58.365.242.375 (XO: 12.365244517)

 - Block number: 2 starting at 2021-07-12T03:00:06.605.309.725 (XO: 20.605312583 - 20.605618584, t=37.2 degC)  Rank: 00083
   TM values: Max =   232  Min =  -268  Ave =    -4.57  Std =    21.49
   Min TM value:-268 at an offset of 109.125 usec -> 2021-07-12T03:00:06.605.418.850 (XO: 20.605421708)
   Max TM value:232 at an offset of 108.838 usec -> 2021-07-12T03:00:06.605.418.562 (XO: 20.605421421)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 111.575 usec -> 2021-07-12T03:00:06.605.421.300 (XO: 20.605424158)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 108.988 usec -> 2021-07-12T03:00:06.605.418.712 (XO: 20.605421571)

 - Block number: 3 starting at 2021-07-12T03:00:18.785.446.725 (XO: 32.785449596 - 32.785755597, t=37.2 degC)  Rank: 00077
   TM values: Max =   236  Min =  -276  Ave =    -4.57  Std =    22.77
   Min TM value:-276 at an offset of 105.688 usec -> 2021-07-12T03:00:18.785.552.412 (XO: 32.785555283)
   Max TM value:236 at an offset of 105.400 usec -> 2021-07-12T03:00:18.785.552.125 (XO: 32.785554996)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 108.088 usec -> 2021-07-12T03:00:18.785.554.812 (XO: 32.785557683)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 105.513 usec -> 2021-07-12T03:00:18.785.552.237 (XO: 32.785555108)
