-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 10 23:03:33 2021 UTC
-- File 20210711_002336_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.511963 to 57.381013
-each separately from 11s to 57s and from 102069us to 709600us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.197059 in 1 packets: 18
      25.392469 in 1 packets: 36
      14.278366 in 1 packets: 54
MIU packet time from 2021-07-11T00:23:50.511.963.000 to 2021-07-11T00:24:36.381.013.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.197059 in 1 packets: 18
      25.392469 in 1 packets: 36
      14.278366 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9408104 to 58.9406563
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999799 in 1 packets: 18
      25.9999129 in 1 packets: 36
      13.9999531 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      1 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      1 in 3 packets: 1, 2, 3
      2 in 52 packets: 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21, ...
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210711_002336_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      254 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4527737 to 13.4530627
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.1970390  -- OK for SL --
   MIU time from 2021-07-11T00:23:50.511.963.000 to 2021-07-11T00:23:50.512.252.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.197059

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.6501017 to 19.6503907
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.3923830  -- OK for SL --
   MIU time from 2021-07-11T00:23:56.709.311.000 to 2021-07-11T00:23:56.709.600.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.392469

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.0427737 to 45.0430627
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2783190  -- OK for SL --
   MIU time from 2021-07-11T00:24:22.102.069.000 to 2021-07-11T00:24:22.102.358.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.278366

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      156 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.3213817 to 59.3216707
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-11T00:24:36.380.724.000 to 2021-07-11T00:24:36.381.013.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.452773700 corrected by 40bit 10MHz counter of 12.940810400 obtained at 1Hz counter of 11. Result: 11.511964989
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-11T00:23:50.511.822.725
   Max 1st sample MIU time among all SL packets: 2021-07-11T00:24:36.380.872.725
   Min 1st sample XO time among all SL packets: 11.511824714
   Max 1st sample XO time among all SL packets: 57.380875382
   TM values from -346 to 434 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-11T00:23:50.511.822.725 (XO: 11.511824714 - 11.512130715, t=37.2 degC)  Rank: 00254
   TM values: Max =   434  Min =  -346  Ave =    -5.19  Std =    55.88
   Min TM value:-346 at an offset of 89.125 usec -> 2021-07-11T00:23:50.511.911.850 (XO: 11.511913839)
   Max TM value:434 at an offset of 85.113 usec -> 2021-07-11T00:23:50.511.907.837 (XO: 11.511909827)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 123.438 usec -> 2021-07-11T00:23:50.511.946.162 (XO: 11.511948152)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 191.113 usec -> 2021-07-11T00:23:50.512.013.838 (XO: 11.512015827)

 - Block number: 2 starting at 2021-07-11T00:23:56.709.170.725 (XO: 17.709173536 - 17.709479537, t=37.2 degC)  Rank: 00080
   TM values: Max =   188  Min =  -227  Ave =    -4.56  Std =    26.49
   Min TM value:-227 at an offset of 173.676 usec -> 2021-07-11T00:23:56.709.344.400 (XO: 17.709347212)
   Max TM value:188 at an offset of 160.138 usec -> 2021-07-11T00:23:56.709.330.863 (XO: 17.709333674)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 111.450 usec -> 2021-07-11T00:23:56.709.282.175 (XO: 17.709284987)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 173.526 usec -> 2021-07-11T00:23:56.709.344.250 (XO: 17.709347062)

 - Block number: 3 starting at 2021-07-11T00:24:22.101.928.725 (XO: 43.101930572 - 43.102236573, t=37.2 degC)  Rank: 00076
   TM values: Max =   202  Min =  -228  Ave =    -4.50  Std =    20.34
   Min TM value:-228 at an offset of 131.213 usec -> 2021-07-11T00:24:22.102.059.937 (XO: 43.102061785)
   Max TM value:202 at an offset of 130.888 usec -> 2021-07-11T00:24:22.102.059.612 (XO: 43.102061460)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 128.750 usec -> 2021-07-11T00:24:22.102.057.475 (XO: 43.102059322)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 131.013 usec -> 2021-07-11T00:24:22.102.059.737 (XO: 43.102061585)

 - Block number: 4 starting at 2021-07-11T00:24:36.380.583.725 (XO: 57.380586381 - 57.380892382, t=37.2 degC)  Rank: 00156
   TM values: Max =   257  Min =  -178  Ave =    -4.50  Std =    34.24
   Min TM value:-178 at an offset of 131.263 usec -> 2021-07-11T00:24:36.380.714.987 (XO: 57.380717644)
   Max TM value:257 at an offset of 109.313 usec -> 2021-07-11T00:24:36.380.693.037 (XO: 57.380695694)
   Max dTM/dt [1/us]:8.31e+02 at an offset of 28.525 usec -> 2021-07-11T00:24:36.380.612.250 (XO: 57.380614906)
   Min dTM/dt [1/us]:-8.16e+02 at an offset of 229.838 usec -> 2021-07-11T00:24:36.380.813.563 (XO: 57.380816219)
