-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 10 20:54:04 2021 UTC
-- File 20210710_164229_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.363136 to 50.544885
-each separately from 6s to 50s and from 123454us to 544885us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.760029 in 1 packets: 18
      18.400236 in 1 packets: 36
      17.020328 in 1 packets: 54
MIU packet time from 2021-07-10T16:42:38.363.136.000 to 2021-07-10T16:43:22.544.885.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.760029 in 1 packets: 18
      18.400236 in 1 packets: 36
      17.020328 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9153724 to 51.9152243
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999697 in 1 packets: 18
      17.9999394 in 1 packets: 36
      16.9999428 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210710_164229_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2785095 to 8.2787985
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7599990  -- OK for SL --
   MIU time from 2021-07-10T16:42:38.363.136.000 to 2021-07-10T16:42:38.363.425.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.760029

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0387975 to 17.0390865
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4001750  -- OK for SL --
   MIU time from 2021-07-10T16:42:47.123.454.000 to 2021-07-10T16:42:47.123.743.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.400236

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4392615 to 35.4395505
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0202710  -- OK for SL --
   MIU time from 2021-07-10T16:43:05.523.979.000 to 2021-07-10T16:43:05.524.268.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.020328

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.4598215 to 52.4601105
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-10T16:43:22.544.596.000 to 2021-07-10T16:43:22.544.885.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.278509500 corrected by 40bit 10MHz counter of 7.915372400 obtained at 1Hz counter of 6. Result: 6.363138298
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-10T16:42:38.362.995.725
   Max 1st sample MIU time among all SL packets: 2021-07-10T16:43:22.544.744.725
   Min 1st sample XO time among all SL packets: 6.362998023
   Max 1st sample XO time among all SL packets: 50.544747723
   TM values from -288 to 254 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-10T16:42:38.362.995.725 (XO: 6.362998023 - 6.363304024, t=37.2 degC)  Rank: 00078
   TM values: Max =   224  Min =  -288  Ave =    -4.45  Std =    23.75
   Min TM value:-288 at an offset of 121.838 usec -> 2021-07-10T16:42:38.363.117.562 (XO: 6.363119861)
   Max TM value:224 at an offset of 121.538 usec -> 2021-07-10T16:42:38.363.117.262 (XO: 6.363119561)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 124.225 usec -> 2021-07-10T16:42:38.363.119.950 (XO: 6.363122248)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 121.625 usec -> 2021-07-10T16:42:38.363.117.350 (XO: 6.363119648)

 - Block number: 2 starting at 2021-07-10T16:42:47.123.313.725 (XO: 15.123315544 - 15.123621545, t=37.2 degC)  Rank: 00081
   TM values: Max =   215  Min =  -256  Ave =    -4.37  Std =    24.39
   Min TM value:-256 at an offset of 108.575 usec -> 2021-07-10T16:42:47.123.422.300 (XO: 15.123424120)
   Max TM value:215 at an offset of 124.413 usec -> 2021-07-10T16:42:47.123.438.137 (XO: 15.123439957)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 110.963 usec -> 2021-07-10T16:42:47.123.424.687 (XO: 15.123426507)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 108.375 usec -> 2021-07-10T16:42:47.123.422.100 (XO: 15.123423920)

 - Block number: 3 starting at 2021-07-10T16:43:05.523.838.725 (XO: 33.523841506 - 33.524147507, t=37.2 degC)  Rank: 00081
   TM values: Max =   254  Min =  -263  Ave =    -4.39  Std =    22.12
   Min TM value:-263 at an offset of 124.125 usec -> 2021-07-10T16:43:05.523.962.850 (XO: 33.523965631)
   Max TM value:254 at an offset of 123.850 usec -> 2021-07-10T16:43:05.523.962.575 (XO: 33.523965356)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 126.500 usec -> 2021-07-10T16:43:05.523.965.225 (XO: 33.523968006)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 123.963 usec -> 2021-07-10T16:43:05.523.962.687 (XO: 33.523965469)

 - Block number: 4 starting at 2021-07-10T16:43:22.544.455.725 (XO: 50.544458722 - 50.544764723, t=37.2 degC)  Rank: 00079
   TM values: Max =   238  Min =  -260  Ave =    -4.47  Std =    22.32
   Min TM value:-260 at an offset of 111.788 usec -> 2021-07-10T16:43:22.544.567.512 (XO: 50.544570510)
   Max TM value:238 at an offset of 109.188 usec -> 2021-07-10T16:43:22.544.564.912 (XO: 50.544567910)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 111.850 usec -> 2021-07-10T16:43:22.544.567.575 (XO: 50.544570572)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 109.288 usec -> 2021-07-10T16:43:22.544.565.012 (XO: 50.544568010)
