-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 10 20:31:48 2021 UTC
-- File 20210710_160946_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.646531 to 23.681360
-each separately from 10s to 23s and from 646531us to 681360us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.034251 in 1 packets: 18
MIU packet time from 2021-07-10T16:09:59.646.531.000 to 2021-07-10T16:10:12.681.360.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.034251 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9136260 to 24.9135819
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999559 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210710_160946_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.5601575 to 12.5604465
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0342070  -- OK for SL --
   MIU time from 2021-07-10T16:09:59.646.531.000 to 2021-07-10T16:09:59.646.820.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.034251

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.5946535 to 25.5949425
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-10T16:10:12.681.071.000 to 2021-07-10T16:10:12.681.360.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.560157500 corrected by 40bit 10MHz counter of 11.913626000 obtained at 1Hz counter of 10. Result: 10.646533634
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-10T16:09:59.646.390.725
   Max 1st sample MIU time among all SL packets: 2021-07-10T16:10:12.681.219.725
   Min 1st sample XO time among all SL packets: 10.646393358
   Max 1st sample XO time among all SL packets: 23.681222641
   TM values from -373 to 343 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-10T16:09:59.646.390.725 (XO: 10.646393358 - 10.646699359, t=37.2 degC)  Rank: 00077
   TM values: Max =   343  Min =  -373  Ave =    -4.45  Std =    66.14
   Min TM value:-373 at an offset of 92.250 usec -> 2021-07-10T16:09:59.646.482.975 (XO: 10.646485608)
   Max TM value:343 at an offset of 89.738 usec -> 2021-07-10T16:09:59.646.480.462 (XO: 10.646483096)
   Max dTM/dt [1/us]:3.05e+03 at an offset of 92.363 usec -> 2021-07-10T16:09:59.646.483.087 (XO: 10.646485721)
   Min dTM/dt [1/us]:-3.03e+03 at an offset of 89.850 usec -> 2021-07-10T16:09:59.646.480.575 (XO: 10.646483208)

 - Block number: 2 starting at 2021-07-10T16:10:12.680.930.725 (XO: 23.680933640 - 23.681239641, t=37.2 degC)  Rank: 00076
   TM values: Max =   136  Min =  -148  Ave =    -4.55  Std =    58.69
   Min TM value:-148 at an offset of 136.050 usec -> 2021-07-10T16:10:12.681.066.775 (XO: 23.681069691)
   Max TM value:136 at an offset of 129.138 usec -> 2021-07-10T16:10:12.681.059.862 (XO: 23.681062778)
   Max dTM/dt [1/us]:1.17e+03 at an offset of 137.138 usec -> 2021-07-10T16:10:12.681.067.862 (XO: 23.681070778)
   Min dTM/dt [1/us]:-1.16e+03 at an offset of 129.188 usec -> 2021-07-10T16:10:12.681.059.912 (XO: 23.681062828)
