-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Nov 10 13:58:48 2021 UTC
-- File 20210710_101941_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.959592 to 69.299035
-each separately from 2s to 69s and from 298746us to 959881us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.449424 in 1 packets: 18
      12.079479 in 1 packets: 36
      14.279471 in 1 packets: 54
      16.529624 in 1 packets: 72
MIU packet time from 2021-07-10T10:19:46.959.592.000 to 2021-07-10T10:20:53.299.035.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.449424 in 1 packets: 18
      12.079479 in 1 packets: 36
      14.279471 in 1 packets: 54
      16.529624 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9228180 to 70.9225917
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999188 in 1 packets: 18
      11.9999594 in 1 packets: 36
      13.9999530 in 1 packets: 54
      16.9999425 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210710_101941_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8824105 to 4.8826995
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.4493430  -- OK for SL --
   MIU time from 2021-07-10T10:19:46.959.592.000 to 2021-07-10T10:19:46.959.881.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.449424

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3320425 to 28.3323315
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0794390  -- OK for SL --
   MIU time from 2021-07-10T10:20:10.409.305.000 to 2021-07-10T10:20:10.409.594.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.079479

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.4117705 to 40.4120595
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2794230  -- OK for SL --
   MIU time from 2021-07-10T10:20:22.489.073.000 to 2021-07-10T10:20:22.489.362.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.279471

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.6914825 to 54.6917715
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5295670  -- OK for SL --
   MIU time from 2021-07-10T10:20:36.768.833.000 to 2021-07-10T10:20:36.769.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.529624

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.2213385 to 71.2216275
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-10T10:20:53.298.746.000 to 2021-07-10T10:20:53.299.035.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.882410500 corrected by 40bit 10MHz counter of 3.922818000 obtained at 1Hz counter of 2. Result: 2.959595763
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-10T10:19:46.959.451.725
   Max 1st sample MIU time among all SL packets: 2021-07-10T10:20:53.298.894.725
   Min 1st sample XO time among all SL packets: 2.959455487
   Max 1st sample XO time among all SL packets: 69.298896541
   TM values from -258 to 238 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-10T10:19:46.959.451.725 (XO: 2.959455487 - 2.959761488, t=37.2 degC)  Rank: 00076
   TM values: Max =   228  Min =  -256  Ave =    -4.48  Std =    21.77
   Min TM value:-256 at an offset of 130.813 usec -> 2021-07-10T10:19:46.959.582.537 (XO: 2.959586300)
   Max TM value:228 at an offset of 128.550 usec -> 2021-07-10T10:19:46.959.580.275 (XO: 2.959584038)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 128.425 usec -> 2021-07-10T10:19:46.959.580.150 (XO: 2.959583913)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 130.625 usec -> 2021-07-10T10:19:46.959.582.350 (XO: 2.959586113)

 - Block number: 2 starting at 2021-07-10T10:20:10.409.164.725 (XO: 26.409166775 - 26.409472776, t=37.2 degC)  Rank: 00080
   TM values: Max =   199  Min =  -245  Ave =    -4.40  Std =    17.84
   Min TM value:-245 at an offset of 113.238 usec -> 2021-07-10T10:20:10.409.277.962 (XO: 26.409280013)
   Max TM value:199 at an offset of 115.875 usec -> 2021-07-10T10:20:10.409.280.600 (XO: 26.409282651)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 115.650 usec -> 2021-07-10T10:20:10.409.280.375 (XO: 26.409282426)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 113.063 usec -> 2021-07-10T10:20:10.409.277.787 (XO: 26.409279838)

 - Block number: 3 starting at 2021-07-10T10:20:22.488.932.725 (XO: 38.488935638 - 38.489241639, t=37.2 degC)  Rank: 00079
   TM values: Max =   212  Min =  -219  Ave =    -4.45  Std =    20.28
   Min TM value:-219 at an offset of 123.488 usec -> 2021-07-10T10:20:22.489.056.212 (XO: 38.489059126)
   Max TM value:212 at an offset of 126.163 usec -> 2021-07-10T10:20:22.489.058.887 (XO: 38.489061801)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 126.013 usec -> 2021-07-10T10:20:22.489.058.737 (XO: 38.489061651)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 123.250 usec -> 2021-07-10T10:20:22.489.055.975 (XO: 38.489058889)

 - Block number: 4 starting at 2021-07-10T10:20:36.768.692.725 (XO: 52.768695715 - 52.769001717, t=37.2 degC)  Rank: 00078
   TM values: Max =   216  Min =  -258  Ave =    -4.38  Std =    27.24
   Min TM value:-258 at an offset of 114.338 usec -> 2021-07-10T10:20:36.768.807.062 (XO: 52.768810053)
   Max TM value:216 at an offset of 114.038 usec -> 2021-07-10T10:20:36.768.806.762 (XO: 52.768809753)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 116.738 usec -> 2021-07-10T10:20:36.768.809.462 (XO: 52.768812453)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 114.138 usec -> 2021-07-10T10:20:36.768.806.862 (XO: 52.768809853)

 - Block number: 5 starting at 2021-07-10T10:20:53.298.605.725 (XO: 69.298607540 - 69.298913541, t=37.2 degC)  Rank: 00083
   TM values: Max =   238  Min =  -240  Ave =    -4.43  Std =    22.97
   Min TM value:-240 at an offset of 128.738 usec -> 2021-07-10T10:20:53.298.734.462 (XO: 69.298736278)
   Max TM value:238 at an offset of 128.488 usec -> 2021-07-10T10:20:53.298.734.212 (XO: 69.298736028)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 126.450 usec -> 2021-07-10T10:20:53.298.732.175 (XO: 69.298733991)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 128.600 usec -> 2021-07-10T10:20:53.298.734.325 (XO: 69.298736141)
