-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 20:53:02 2021 UTC
-- File 20210709_085050_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.475097 to 26.965576
-each separately from 8s to 26s and from 475097us to 965576us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.489901 in 1 packets: 18
MIU packet time from 2021-07-09T08:51:01.475.097.000 to 2021-07-09T08:51:19.965.576.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.489901 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8229938 to 27.8229323
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999385 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210709_085050_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2980916 to 10.2983806
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4898390  -- OK for SL --
   MIU time from 2021-07-09T08:51:01.475.097.000 to 2021-07-09T08:51:01.475.386.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.489901

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.7882196 to 28.7885086
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-09T08:51:19.965.287.000 to 2021-07-09T08:51:19.965.576.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.298091600 corrected by 40bit 10MHz counter of 9.822993800 obtained at 1Hz counter of 8. Result: 8.475099415
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-07-09T08:51:01.474.956.725
   Max 1st sample MIU time among all SL packets: 2021-07-09T08:51:19.965.435.725
   Min 1st sample XO time among all SL packets: 8.474959140
   Max 1st sample XO time among all SL packets: 26.965439307
   TM values from -260 to 251 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-09T08:51:01.474.956.725 (XO: 8.474959140 - 8.475265141, t=37.2 degC)  Rank: 00080
   TM values: Max =   251  Min =  -260  Ave =    -4.46  Std =    30.85
   Min TM value:-260 at an offset of 174.013 usec -> 2021-07-09T08:51:01.475.130.738 (XO: 8.475133153)
   Max TM value:251 at an offset of 119.688 usec -> 2021-07-09T08:51:01.475.076.412 (XO: 8.475078828)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 119.563 usec -> 2021-07-09T08:51:01.475.076.287 (XO: 8.475078703)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 173.826 usec -> 2021-07-09T08:51:01.475.130.550 (XO: 8.475132965)

 - Block number: 2 starting at 2021-07-09T08:51:19.965.146.725 (XO: 26.965150307 - 26.965456308, t=37.2 degC)  Rank: 00082
   TM values: Max =   228  Min =  -236  Ave =    -4.47  Std =    19.80
   Min TM value:-236 at an offset of 116.050 usec -> 2021-07-09T08:51:19.965.262.775 (XO: 26.965266357)
   Max TM value:228 at an offset of 118.788 usec -> 2021-07-09T08:51:19.965.265.512 (XO: 26.965269094)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 118.625 usec -> 2021-07-09T08:51:19.965.265.350 (XO: 26.965268932)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 115.863 usec -> 2021-07-09T08:51:19.965.262.587 (XO: 26.965266169)
