-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 20:24:47 2021 UTC
-- File 20210709_083010_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.425662 to 15.241250
-each separately from 0s to 15s and from 240961us to 425951us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.815010 in 1 packets: 18
MIU packet time from 2021-07-09T08:30:13.425.662.000 to 2021-07-09T08:30:28.241.250.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.815010 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8979465 to 16.8978952
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999487 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210709_083010_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.3236094 to 2.3238984
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8149590  -- OK for SL --
   MIU time from 2021-07-09T08:30:13.425.662.000 to 2021-07-09T08:30:13.425.951.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.815010

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1388574 to 17.1391464
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-09T08:30:28.240.961.000 to 2021-07-09T08:30:28.241.250.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.323609400 corrected by 40bit 10MHz counter of 1.897946500 obtained at 1Hz counter of 0. Result: 0.425664390
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-09T08:30:13.425.521.725
   Max 1st sample MIU time among all SL packets: 2021-07-09T08:30:28.241.109.725
   Min 1st sample XO time among all SL packets: 0.425524114
   Max 1st sample XO time among all SL packets: 15.241111769
   TM values from -278 to 253 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-09T08:30:13.425.521.725 (XO: 0.425524114 - 0.425830115, t=37.2 degC)  Rank: 00078
   TM values: Max =   244  Min =  -275  Ave =    -4.53  Std =    25.91
   Min TM value:-275 at an offset of 117.013 usec -> 2021-07-09T08:30:13.425.638.737 (XO: 0.425641127)
   Max TM value:244 at an offset of 116.663 usec -> 2021-07-09T08:30:13.425.638.387 (XO: 0.425640777)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 119.488 usec -> 2021-07-09T08:30:13.425.641.212 (XO: 0.425643602)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 116.775 usec -> 2021-07-09T08:30:13.425.638.500 (XO: 0.425640890)

 - Block number: 2 starting at 2021-07-09T08:30:28.240.820.725 (XO: 15.240822768 - 15.241128769, t=37.2 degC)  Rank: 00085
   TM values: Max =   253  Min =  -278  Ave =    -4.56  Std =    27.58
   Min TM value:-278 at an offset of 93.400 usec -> 2021-07-09T08:30:28.240.914.125 (XO: 15.240916168)
   Max TM value:253 at an offset of 105.588 usec -> 2021-07-09T08:30:28.240.926.312 (XO: 15.240928356)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 108.413 usec -> 2021-07-09T08:30:28.240.929.137 (XO: 15.240931181)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 105.700 usec -> 2021-07-09T08:30:28.240.926.425 (XO: 15.240928468)
