-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 14:30:28 2021 UTC
-- File 20210709_022406_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.013341 to 28.273869
-each separately from 0s to 28s and from 13341us to 273869us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.259950 in 1 packets: 18
MIU packet time from 2021-07-09T02:24:09.013.341.000 to 2021-07-09T02:24:37.273.869.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.259950 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8352202 to 29.8351258
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999056 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      3 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      3 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210709_022406_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8485628 to 1.8488518
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.2598550  -- OK for SL --
   MIU time from 2021-07-09T02:24:09.013.341.000 to 2021-07-09T02:24:09.013.630.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.259950

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.1087068 to 30.1089958
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-09T02:24:37.273.580.000 to 2021-07-09T02:24:37.273.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.848562800 corrected by 40bit 10MHz counter of 1.835220200 obtained at 1Hz counter of 0. Result: 0.013342644
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-09T02:24:09.013.200.725
   Max 1st sample MIU time among all SL packets: 2021-07-09T02:24:37.273.728.725
   Min 1st sample XO time among all SL packets: 0.013202369
   Max 1st sample XO time among all SL packets: 28.273730656
   TM values from -269 to 215 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-09T02:24:09.013.200.725 (XO: 0.013202369 - 0.013508370, t=37.2 degC)  Rank: 00083
   TM values: Max =   212  Min =  -269  Ave =    -4.52  Std =    19.75
   Min TM value:-269 at an offset of 122.713 usec -> 2021-07-09T02:24:09.013.323.437 (XO: 0.013325081)
   Max TM value:212 at an offset of 125.363 usec -> 2021-07-09T02:24:09.013.326.087 (XO: 0.013327731)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 125.188 usec -> 2021-07-09T02:24:09.013.325.912 (XO: 0.013327556)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 122.588 usec -> 2021-07-09T02:24:09.013.323.312 (XO: 0.013324956)

 - Block number: 2 starting at 2021-07-09T02:24:37.273.439.725 (XO: 28.273441655 - 28.273747656, t=37.2 degC)  Rank: 00079
   TM values: Max =   215  Min =  -235  Ave =    -4.59  Std =    19.50
   Min TM value:-235 at an offset of 121.863 usec -> 2021-07-09T02:24:37.273.561.587 (XO: 28.273563518)
   Max TM value:215 at an offset of 124.525 usec -> 2021-07-09T02:24:37.273.564.250 (XO: 28.273566180)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 124.350 usec -> 2021-07-09T02:24:37.273.564.075 (XO: 28.273566005)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 121.688 usec -> 2021-07-09T02:24:37.273.561.412 (XO: 28.273563343)
