-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 13:49:42 2021 UTC
-- File 20210709_020320_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.591794 to 19.782111
-each separately from 11s to 19s and from 591794us to 782111us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.189739 in 1 packets: 18
MIU packet time from 2021-07-09T02:03:34.591.794.000 to 2021-07-09T02:03:42.782.111.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.189739 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.9329861 to 20.9329590
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999729 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210709_020320_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.5247812 to 13.5250702
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1897110  -- OK for SL --
   MIU time from 2021-07-09T02:03:34.591.794.000 to 2021-07-09T02:03:34.592.083.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.189739

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7147812 to 21.7150702
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-09T02:03:42.781.822.000 to 2021-07-09T02:03:42.782.111.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.524781200 corrected by 40bit 10MHz counter of 12.932986100 obtained at 1Hz counter of 11. Result: 11.591797053
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-09T02:03:34.591.653.725
   Max 1st sample MIU time among all SL packets: 2021-07-09T02:03:42.781.970.725
   Min 1st sample XO time among all SL packets: 11.591656777
   Max 1st sample XO time among all SL packets: 19.781973506
   TM values from -260 to 215 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-09T02:03:34.591.653.725 (XO: 11.591656777 - 11.591962778, t=37.2 degC)  Rank: 00085
   TM values: Max =   215  Min =  -257  Ave =    -4.54  Std =    20.36
   Min TM value:-257 at an offset of 108.200 usec -> 2021-07-09T02:03:34.591.761.925 (XO: 11.591764978)
   Max TM value:215 at an offset of 107.900 usec -> 2021-07-09T02:03:34.591.761.625 (XO: 11.591764678)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 110.625 usec -> 2021-07-09T02:03:34.591.764.350 (XO: 11.591767403)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 108.038 usec -> 2021-07-09T02:03:34.591.761.762 (XO: 11.591764815)

 - Block number: 2 starting at 2021-07-09T02:03:42.781.681.725 (XO: 19.781684505 - 19.781990506, t=37.2 degC)  Rank: 00081
   TM values: Max =   208  Min =  -260  Ave =    -4.55  Std =    20.90
   Min TM value:-260 at an offset of 123.425 usec -> 2021-07-09T02:03:42.781.805.150 (XO: 19.781807930)
   Max TM value:208 at an offset of 123.125 usec -> 2021-07-09T02:03:42.781.804.850 (XO: 19.781807630)
   Max dTM/dt [1/us]:1.47e+03 at an offset of 125.938 usec -> 2021-07-09T02:03:42.781.807.662 (XO: 19.781810442)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 123.263 usec -> 2021-07-09T02:03:42.781.804.987 (XO: 19.781807767)
