-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Nov  9 12:26:08 2021 UTC
-- File 20210709_011644_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.189333 to 49.768965
-each separately from 2s to 49s and from 189333us to 768965us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.389373 in 1 packets: 18
      11.179620 in 1 packets: 36
      13.009483 in 1 packets: 54
MIU packet time from 2021-07-09T01:16:49.189.333.000 to 2021-07-09T01:17:36.768.965.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.389373 in 1 packets: 18
      11.179620 in 1 packets: 36
      13.009483 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8971167 to 50.8969579
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999222 in 1 packets: 18
      10.9999628 in 1 packets: 36
      12.9999562 in 1 packets: 54
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210709_011644_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0864507 to 4.0867397
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3892950  -- OK for SL --
   MIU time from 2021-07-09T01:16:49.189.333.000 to 2021-07-09T01:16:49.189.622.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.389373

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4760347 to 27.4763237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1795830  -- OK for SL --
   MIU time from 2021-07-09T01:17:12.578.995.000 to 2021-07-09T01:17:12.579.284.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.179620

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.6559067 to 38.6561957
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0094390  -- OK for SL --
   MIU time from 2021-07-09T01:17:23.758.904.000 to 2021-07-09T01:17:23.759.193.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.009483

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.6656347 to 51.6659237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-09T01:17:36.768.676.000 to 2021-07-09T01:17:36.768.965.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.086450700 corrected by 40bit 10MHz counter of 3.897116700 obtained at 1Hz counter of 2. Result: 2.189334644
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-07-09T01:16:49.189.192.725
   Max 1st sample MIU time among all SL packets: 2021-07-09T01:17:36.768.824.725
   Min 1st sample XO time among all SL packets: 2.189194368
   Max 1st sample XO time among all SL packets: 49.768828216
   TM values from -254 to 226 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-09T01:16:49.189.192.725 (XO: 2.189194368 - 2.189500369, t=37.2 degC)  Rank: 00077
   TM values: Max =   210  Min =  -245  Ave =    -4.51  Std =    22.42
   Min TM value:-245 at an offset of 115.425 usec -> 2021-07-09T01:16:49.189.308.150 (XO: 2.189309794)
   Max TM value:210 at an offset of 118.113 usec -> 2021-07-09T01:16:49.189.310.837 (XO: 2.189312481)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 118.025 usec -> 2021-07-09T01:16:49.189.310.750 (XO: 2.189312394)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 115.325 usec -> 2021-07-09T01:16:49.189.308.050 (XO: 2.189309694)

 - Block number: 2 starting at 2021-07-09T01:17:12.578.854.725 (XO: 25.578857435 - 25.579163436, t=37.2 degC)  Rank: 00079
   TM values: Max =   219  Min =  -254  Ave =    -4.53  Std =    23.84
   Min TM value:-254 at an offset of 127.763 usec -> 2021-07-09T01:17:12.578.982.487 (XO: 25.578985198)
   Max TM value:219 at an offset of 127.463 usec -> 2021-07-09T01:17:12.578.982.187 (XO: 25.578984898)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 130.175 usec -> 2021-07-09T01:17:12.578.984.900 (XO: 25.578987611)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 127.550 usec -> 2021-07-09T01:17:12.578.982.275 (XO: 25.578984986)

 - Block number: 3 starting at 2021-07-09T01:17:23.758.763.725 (XO: 36.758767305 - 36.759073306, t=37.2 degC)  Rank: 00084
   TM values: Max =   218  Min =  -246  Ave =    -4.55  Std =    20.09
   Min TM value:-246 at an offset of 100.525 usec -> 2021-07-09T01:17:23.758.864.250 (XO: 36.758867830)
   Max TM value:218 at an offset of 116.025 usec -> 2021-07-09T01:17:23.758.879.750 (XO: 36.758883330)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 115.863 usec -> 2021-07-09T01:17:23.758.879.587 (XO: 36.758883168)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 113.188 usec -> 2021-07-09T01:17:23.758.876.912 (XO: 36.758880493)

 - Block number: 4 starting at 2021-07-09T01:17:36.768.535.725 (XO: 49.768539215 - 49.768845216, t=37.2 degC)  Rank: 00080
   TM values: Max =   226  Min =  -226  Ave =    -4.45  Std =    26.92
   Min TM value:-226 at an offset of 113.050 usec -> 2021-07-09T01:17:36.768.648.775 (XO: 49.768652265)
   Max TM value:226 at an offset of 115.763 usec -> 2021-07-09T01:17:36.768.651.487 (XO: 49.768654978)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 115.650 usec -> 2021-07-09T01:17:36.768.651.375 (XO: 49.768654865)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 112.888 usec -> 2021-07-09T01:17:36.768.648.612 (XO: 49.768652103)
