-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Oct 29 10:05:37 2021 UTC
-- File 20210702_104255_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.989191 to 18.349453
-each separately from 3s to 18s and from 349164us to 989480us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.359684 in 1 packets: 18
MIU packet time from 2021-07-02T10:42:57.989.191.000 to 2021-07-02T10:43:12.349.453.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.359684 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8813437 to 19.8812899
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999462 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210702_104255_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.8705347 to 5.8708237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3596310  -- OK for SL --
   MIU time from 2021-07-02T10:42:57.989.191.000 to 2021-07-02T10:42:57.989.480.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.359684

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2304547 to 20.2307437
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-02T10:43:12.349.164.000 to 2021-07-02T10:43:12.349.453.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.870534700 corrected by 40bit 10MHz counter of 4.881343700 obtained at 1Hz counter of 3. Result: 3.989194561
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-02T10:42:57.989.050.725
   Max 1st sample MIU time among all SL packets: 2021-07-02T10:43:12.349.312.725
   Min 1st sample XO time among all SL packets: 3.989054286
   Max 1st sample XO time among all SL packets: 18.349314783
   TM values from -258 to 216 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-02T10:42:57.989.050.725 (XO: 3.989054286 - 3.989360287, t=35.6 degC)  Rank: 00084
   TM values: Max =   216  Min =  -245  Ave =    -4.52  Std =    20.49
   Min TM value:-245 at an offset of 115.363 usec -> 2021-07-02T10:42:57.989.166.087 (XO: 3.989169649)
   Max TM value:216 at an offset of 130.700 usec -> 2021-07-02T10:42:57.989.181.425 (XO: 3.989184986)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 130.550 usec -> 2021-07-02T10:42:57.989.181.275 (XO: 3.989184836)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 127.750 usec -> 2021-07-02T10:42:57.989.178.475 (XO: 3.989182036)

 - Block number: 2 starting at 2021-07-02T10:43:12.349.023.725 (XO: 18.349025781 - 18.349331783, t=35.6 degC)  Rank: 00080
   TM values: Max =   210  Min =  -258  Ave =    -4.47  Std =    21.69
   Min TM value:-258 at an offset of 128.725 usec -> 2021-07-02T10:43:12.349.152.450 (XO: 18.349154507)
   Max TM value:210 at an offset of 131.363 usec -> 2021-07-02T10:43:12.349.155.087 (XO: 18.349157144)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 131.225 usec -> 2021-07-02T10:43:12.349.154.950 (XO: 18.349157007)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 128.525 usec -> 2021-07-02T10:43:12.349.152.250 (XO: 18.349154307)
