-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  2 09:07:21 2021 UTC
-- File 20210701_221037_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.958827 to 18.188845
-each separately from 13s to 18s and from 188556us to 959116us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.229440 in 1 packets: 18
MIU packet time from 2021-07-01T22:10:50.958.827.000 to 2021-07-01T22:10:55.188.845.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.229440 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.2596245 to 20.2596063
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999818 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210701_221037_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2184512 to 16.2187402
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.2294230  -- OK for SL --
   MIU time from 2021-07-01T22:10:50.958.827.000 to 2021-07-01T22:10:50.959.116.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.229440

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.4481632 to 20.4484522
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-01T22:10:55.188.556.000 to 2021-07-01T22:10:55.188.845.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 16.218451200 corrected by 40bit 10MHz counter of 15.259624500 obtained at 1Hz counter of 13. Result: 13.958830248
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2021-07-01T22:10:50.958.686.725
   Max 1st sample MIU time among all SL packets: 2021-07-01T22:10:55.188.704.725
   Min 1st sample XO time among all SL packets: 13.958689972
   Max 1st sample XO time among all SL packets: 18.188706323
   TM values from -289 to 255 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-01T22:10:50.958.686.725 (XO: 13.958689972 - 13.958995973, t=35.6 degC)  Rank: 00080
   TM values: Max =   255  Min =  -284  Ave =    -4.38  Std =    22.81
   Min TM value:-284 at an offset of 112.038 usec -> 2021-07-01T22:10:50.958.798.762 (XO: 13.958802010)
   Max TM value:255 at an offset of 114.688 usec -> 2021-07-01T22:10:50.958.801.412 (XO: 13.958804660)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 114.550 usec -> 2021-07-01T22:10:50.958.801.275 (XO: 13.958804523)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 111.838 usec -> 2021-07-01T22:10:50.958.798.562 (XO: 13.958801810)

 - Block number: 2 starting at 2021-07-01T22:10:55.188.415.725 (XO: 18.188417322 - 18.188723323, t=35.6 degC)  Rank: 00082
   TM values: Max =   251  Min =  -289  Ave =    -4.33  Std =    33.84
   Min TM value:-289 at an offset of 167.526 usec -> 2021-07-01T22:10:55.188.583.250 (XO: 18.188584848)
   Max TM value:251 at an offset of 165.288 usec -> 2021-07-01T22:10:55.188.581.013 (XO: 18.188582610)
   Max dTM/dt [1/us]:2.35e+03 at an offset of 165.151 usec -> 2021-07-01T22:10:55.188.580.875 (XO: 18.188582473)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 208.563 usec -> 2021-07-01T22:10:55.188.624.288 (XO: 18.188625885)
