-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  2 08:55:35 2021 UTC
-- File 20210701_204518_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.048739 to 91.211514
-each separately from 14s to 91s and from 48739us to 871212us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.270559 in 1 packets: 18
      14.040001 in 1 packets: 36
      9.040064 in 1 packets: 54
      19.470404 in 1 packets: 72
      8.439823 in 1 packets: 90
      12.899901 in 1 packets: 108
MIU packet time from 2021-07-01T20:45:31.048.739.000 to 2021-07-01T20:46:48.211.514.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.270559 in 1 packets: 18
      14.040001 in 1 packets: 36
      9.040064 in 1 packets: 54
      19.470404 in 1 packets: 72
      8.439823 in 1 packets: 90
      12.899901 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7640126 to 92.7637351
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999532 in 1 packets: 18
      13.9999495 in 1 packets: 36
      8.9999676 in 2 packets: 54, 90
      18.9999315 in 1 packets: 72
      12.9999531 in 1 packets: 108
Temperature FPGA: 
      05 -> 48.5 degC in 108 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C0 -> 35.6 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 72 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210701_204518_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.8127531 to 15.8130421
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2705110  -- OK for SL --
   MIU time from 2021-07-01T20:45:31.048.739.000 to 2021-07-01T20:45:31.049.028.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.270559

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0835531 to 29.0838421
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0399510  -- OK for SL --
   MIU time from 2021-07-01T20:45:44.319.587.000 to 2021-07-01T20:45:44.319.876.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.040001

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.1237931 to 43.1240821
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0400310  -- OK for SL --
   MIU time from 2021-07-01T20:45:58.359.877.000 to 2021-07-01T20:45:58.360.166.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.040064

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.1641131 to 52.1644021
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4703350  -- OK for SL --
   MIU time from 2021-07-01T20:46:07.400.230.000 to 2021-07-01T20:46:07.400.519.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.470404

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.6347371 to 71.6350261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.4397910  -- OK for SL --
   MIU time from 2021-07-01T20:46:26.870.923.000 to 2021-07-01T20:46:26.871.212.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.439823

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.0748171 to 80.0751061
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8998550  -- OK for SL --
   MIU time from 2021-07-01T20:46:35.311.035.000 to 2021-07-01T20:46:35.311.324.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.899901

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 92.9749611 to 92.9752501
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-01T20:46:48.211.225.000 to 2021-07-01T20:46:48.211.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.812753100 corrected by 40bit 10MHz counter of 15.764012600 obtained at 1Hz counter of 14. Result: 14.048740675
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-01T20:45:31.048.598.725
   Max 1st sample MIU time among all SL packets: 2021-07-01T20:46:48.211.373.725
   Min 1st sample XO time among all SL packets: 14.048600400
   Max 1st sample XO time among all SL packets: 91.211375486
   TM values from -303 to 277 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-01T20:45:31.048.598.725 (XO: 14.048600400 - 14.048906401, t=35.6 degC)  Rank: 00077
   TM values: Max =   259  Min =  -249  Ave =    -4.44  Std =    29.79
   Min TM value:-249 at an offset of 100.963 usec -> 2021-07-01T20:45:31.048.699.687 (XO: 14.048701363)
   Max TM value:259 at an offset of 116.325 usec -> 2021-07-01T20:45:31.048.715.050 (XO: 14.048716725)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 116.125 usec -> 2021-07-01T20:45:31.048.714.850 (XO: 14.048716525)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 113.488 usec -> 2021-07-01T20:45:31.048.712.212 (XO: 14.048713888)

 - Block number: 2 starting at 2021-07-01T20:45:44.319.446.725 (XO: 27.319448175 - 27.319754176, t=35.6 degC)  Rank: 00075
   TM values: Max =   255  Min =  -265  Ave =    -4.40  Std =    25.43
   Min TM value:-265 at an offset of 45.200 usec -> 2021-07-01T20:45:44.319.491.925 (XO: 27.319493375)
   Max TM value:255 at an offset of 47.825 usec -> 2021-07-01T20:45:44.319.494.550 (XO: 27.319496000)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 47.688 usec -> 2021-07-01T20:45:44.319.494.412 (XO: 27.319495863)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 44.988 usec -> 2021-07-01T20:45:44.319.491.712 (XO: 27.319493163)

 - Block number: 3 starting at 2021-07-01T20:45:58.359.736.725 (XO: 41.359738820 - 41.360044821, t=35.6 degC)  Rank: 00081
   TM values: Max =   277  Min =  -277  Ave =    -4.42  Std =    27.05
   Min TM value:-277 at an offset of 42.275 usec -> 2021-07-01T20:45:58.359.779.000 (XO: 41.359781095)
   Max TM value:277 at an offset of 44.913 usec -> 2021-07-01T20:45:58.359.781.637 (XO: 41.359783733)
   Max dTM/dt [1/us]:2.53e+03 at an offset of 44.788 usec -> 2021-07-01T20:45:58.359.781.512 (XO: 41.359783608)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 42.088 usec -> 2021-07-01T20:45:58.359.778.812 (XO: 41.359780908)

 - Block number: 4 starting at 2021-07-01T20:46:07.400.089.725 (XO: 50.400091365 - 50.400397366, t=35.6 degC)  Rank: 00073
   TM values: Max =   238  Min =  -265  Ave =    -4.48  Std =    31.53
   Min TM value:-265 at an offset of 129.450 usec -> 2021-07-01T20:46:07.400.219.175 (XO: 50.400220816)
   Max TM value:238 at an offset of 132.175 usec -> 2021-07-01T20:46:07.400.221.900 (XO: 50.400223541)
   Max dTM/dt [1/us]:2.62e+03 at an offset of 132.013 usec -> 2021-07-01T20:46:07.400.221.737 (XO: 50.400223378)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 59.588 usec -> 2021-07-01T20:46:07.400.149.312 (XO: 50.400150953)

 - Block number: 5 starting at 2021-07-01T20:46:26.870.782.725 (XO: 69.870785560 - 69.871091561, t=35.6 degC)  Rank: 00080
   TM values: Max =   241  Min =  -303  Ave =    -4.40  Std =    28.37
   Min TM value:-303 at an offset of 113.163 usec -> 2021-07-01T20:46:26.870.895.887 (XO: 69.870898723)
   Max TM value:241 at an offset of 112.875 usec -> 2021-07-01T20:46:26.870.895.600 (XO: 69.870898435)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 115.738 usec -> 2021-07-01T20:46:26.870.898.462 (XO: 69.870901298)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 113.050 usec -> 2021-07-01T20:46:26.870.895.775 (XO: 69.870898610)

 - Block number: 6 starting at 2021-07-01T20:46:35.310.894.725 (XO: 78.310895944 - 78.311201945, t=35.6 degC)  Rank: 00079
   TM values: Max =   239  Min =  -278  Ave =    -4.49  Std =    27.87
   Min TM value:-278 at an offset of 101.813 usec -> 2021-07-01T20:46:35.310.996.537 (XO: 78.310997757)
   Max TM value:239 at an offset of 101.463 usec -> 2021-07-01T20:46:35.310.996.187 (XO: 78.310997407)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 104.288 usec -> 2021-07-01T20:46:35.310.999.012 (XO: 78.311000232)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 89.238 usec -> 2021-07-01T20:46:35.310.983.962 (XO: 78.310985182)

 - Block number: 7 starting at 2021-07-01T20:46:48.211.084.725 (XO: 91.211086485 - 91.211392486, t=35.6 degC)  Rank: 00082
   TM values: Max =   243  Min =  -252  Ave =    -4.45  Std =    23.17
   Min TM value:-252 at an offset of 115.850 usec -> 2021-07-01T20:46:48.211.200.575 (XO: 91.211202335)
   Max TM value:243 at an offset of 118.575 usec -> 2021-07-01T20:46:48.211.203.300 (XO: 91.211205060)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 118.400 usec -> 2021-07-01T20:46:48.211.203.125 (XO: 91.211204885)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 115.738 usec -> 2021-07-01T20:46:48.211.200.462 (XO: 91.211202223)
