-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  2 05:15:43 2021 UTC
-- File 20210701_124343_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.621333 to 6.621622
-each separately from 6s to 6s and from 621333us to 621622us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-07-01T12:43:48.621.333.000 to 2021-07-01T12:43:48.621.622.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9190812 to 7.9190812
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20210701_124343_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5404148 to 8.5407038
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-01T12:43:48.621.333.000 to 2021-07-01T12:43:48.621.622.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.540414800 corrected by 40bit 10MHz counter of 7.919081200 obtained at 1Hz counter of 6. Result: 6.621335837
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-01T12:43:48.621.192.725
   Max 1st sample MIU time among all SL packets: 2021-07-01T12:43:48.621.481.725
   Min 1st sample XO time among all SL packets: 6.621195561
   Max 1st sample XO time among all SL packets: 6.621484562
   TM values from -250 to 239 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-01T12:43:48.621.192.725 (XO: 6.621195561 - 6.621501562, t=35.6 degC)  Rank: 00078
   TM values: Max =   239  Min =  -250  Ave =    -4.41  Std =    23.11
   Min TM value:-250 at an offset of 106.838 usec -> 2021-07-01T12:43:48.621.299.562 (XO: 6.621302399)
   Max TM value:239 at an offset of 109.475 usec -> 2021-07-01T12:43:48.621.302.200 (XO: 6.621305037)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 109.288 usec -> 2021-07-01T12:43:48.621.302.012 (XO: 6.621304849)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 138.100 usec -> 2021-07-01T12:43:48.621.330.825 (XO: 6.621333662)
