-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  2 03:17:12 2021 UTC
-- File 20210701_101233_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.338279 to 29.992706
-each separately from 13s to 29s and from 338279us to 992706us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.653849 in 1 packets: 18
MIU packet time from 2021-07-01T10:12:45.338.279.000 to 2021-07-01T10:13:01.992.706.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.653849 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9420811 to 30.9420247
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999436 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C0 -> 35.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210701_101233_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.2803612 to 15.2806502
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6537910  -- OK for SL --
   MIU time from 2021-07-01T10:12:45.338.279.000 to 2021-07-01T10:12:45.338.568.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.653849

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9344412 to 31.9347302
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-01T10:13:01.992.417.000 to 2021-07-01T10:13:01.992.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.280361200 corrected by 40bit 10MHz counter of 14.942081100 obtained at 1Hz counter of 13. Result: 13.338281284
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-01T10:12:45.338.138.725
   Max 1st sample MIU time among all SL packets: 2021-07-01T10:13:01.992.565.725
   Min 1st sample XO time among all SL packets: 13.338141008
   Max 1st sample XO time among all SL packets: 29.992568798
   TM values from -264 to 245 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-01T10:12:45.338.138.725 (XO: 13.338141008 - 13.338447010, t=34.0 degC)  Rank: 00087
   TM values: Max =   245  Min =  -256  Ave =    -4.46  Std =    21.43
   Min TM value:-256 at an offset of 113.513 usec -> 2021-07-01T10:12:45.338.252.237 (XO: 13.338254521)
   Max TM value:245 at an offset of 116.188 usec -> 2021-07-01T10:12:45.338.254.912 (XO: 13.338257196)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 116.050 usec -> 2021-07-01T10:12:45.338.254.775 (XO: 13.338257059)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 113.325 usec -> 2021-07-01T10:12:45.338.252.050 (XO: 13.338254334)

 - Block number: 2 starting at 2021-07-01T10:13:01.992.276.725 (XO: 29.992279797 - 29.992585798, t=35.6 degC)  Rank: 00084
   TM values: Max =   202  Min =  -264  Ave =    -4.39  Std =    18.08
   Min TM value:-264 at an offset of 128.950 usec -> 2021-07-01T10:13:01.992.405.675 (XO: 29.992408748)
   Max TM value:202 at an offset of 128.663 usec -> 2021-07-01T10:13:01.992.405.387 (XO: 29.992408460)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 126.500 usec -> 2021-07-01T10:13:01.992.403.225 (XO: 29.992406298)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 128.763 usec -> 2021-07-01T10:13:01.992.405.487 (XO: 29.992408560)
