-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Aug  2 00:42:19 2021 UTC
-- File 20210701_074917_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.084240 to 23.233945
-each separately from 3s to 23s and from 84240us to 233945us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.149127 in 1 packets: 18
MIU packet time from 2021-07-01T07:49:19.084.240.000 to 2021-07-01T07:49:39.233.945.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.149127 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7372467 to 24.7371754
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999287 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210701_074917_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8214881 to 4.8217771
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1490550  -- OK for SL --
   MIU time from 2021-07-01T07:49:19.084.240.000 to 2021-07-01T07:49:19.084.529.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.149127

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.9708321 to 24.9711211
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-01T07:49:39.233.656.000 to 2021-07-01T07:49:39.233.945.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.821488100 corrected by 40bit 10MHz counter of 4.737246700 obtained at 1Hz counter of 3. Result: 3.084241703
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-07-01T07:49:19.084.099.725
   Max 1st sample MIU time among all SL packets: 2021-07-01T07:49:39.233.804.725
   Min 1st sample XO time among all SL packets: 3.084101428
   Max 1st sample XO time among all SL packets: 23.233806267
   TM values from -263 to 245 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-01T07:49:19.084.099.725 (XO: 3.084101428 - 3.084407429, t=35.6 degC)  Rank: 00080
   TM values: Max =   245  Min =  -229  Ave =    -4.40  Std =    18.84
   Min TM value:-229 at an offset of 86.913 usec -> 2021-07-01T07:49:19.084.186.637 (XO: 3.084188341)
   Max TM value:245 at an offset of 89.638 usec -> 2021-07-01T07:49:19.084.189.362 (XO: 3.084191066)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 89.513 usec -> 2021-07-01T07:49:19.084.189.237 (XO: 3.084190941)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 86.738 usec -> 2021-07-01T07:49:19.084.186.462 (XO: 3.084188166)

 - Block number: 2 starting at 2021-07-01T07:49:39.233.515.725 (XO: 23.233517266 - 23.233823267, t=35.6 degC)  Rank: 00092
   TM values: Max =   243  Min =  -263  Ave =    -4.34  Std =    21.74
   Min TM value:-263 at an offset of 125.175 usec -> 2021-07-01T07:49:39.233.640.900 (XO: 23.233642441)
   Max TM value:243 at an offset of 122.575 usec -> 2021-07-01T07:49:39.233.638.300 (XO: 23.233639841)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 125.288 usec -> 2021-07-01T07:49:39.233.641.012 (XO: 23.233642554)
   Min dTM/dt [1/us]:-2.61e+03 at an offset of 122.700 usec -> 2021-07-01T07:49:39.233.638.425 (XO: 23.233639966)
