-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Aug  1 09:02:11 2021 UTC
-- File 20210701_025037_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.230660 to 46.031487
-each separately from 2s to 46s and from 31198us to 641218us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.409980 in 1 packets: 18
      10.779734 in 1 packets: 36
      10.609957 in 1 packets: 54
MIU packet time from 2021-07-01T02:50:38.230.660.000 to 2021-07-01T02:51:22.031.487.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.409980 in 1 packets: 18
      10.779734 in 1 packets: 36
      10.609957 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9937666 to 47.9936118
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999226 in 1 packets: 18
      10.9999613 in 2 packets: 36, 54
Temperature FPGA: 
      07 -> 45.9 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C0 -> 35.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210701_025037_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2244270 to 4.2247160
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.4099030  -- OK for SL --
   MIU time from 2021-07-01T02:50:38.230.660.000 to 2021-07-01T02:50:38.230.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.409980

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6346190 to 26.6349080
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7796950  -- OK for SL --
   MIU time from 2021-07-01T02:51:00.640.929.000 to 2021-07-01T02:51:00.641.218.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.779734

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.4146030 to 37.4148920
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6099190  -- OK for SL --
   MIU time from 2021-07-01T02:51:11.420.952.000 to 2021-07-01T02:51:11.421.241.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.609957

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.0248110 to 48.0251000
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-07-01T02:51:22.031.198.000 to 2021-07-01T02:51:22.031.487.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.224427000 corrected by 40bit 10MHz counter of 3.993766600 obtained at 1Hz counter of 2. Result: 2.230661207
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-07-01T02:50:38.230.519.725
   Max 1st sample MIU time among all SL packets: 2021-07-01T02:51:22.031.346.725
   Min 1st sample XO time among all SL packets: 2.230520932
   Max 1st sample XO time among all SL packets: 46.031348035
   TM values from -257 to 223 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-07-01T02:50:38.230.519.725 (XO: 2.230520932 - 2.230826933, t=35.6 degC)  Rank: 00082
   TM values: Max =   196  Min =  -233  Ave =    -4.50  Std =    20.20
   Min TM value:-233 at an offset of 110.825 usec -> 2021-07-01T02:50:38.230.630.550 (XO: 2.230631757)
   Max TM value:196 at an offset of 126.500 usec -> 2021-07-01T02:50:38.230.646.225 (XO: 2.230647432)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 126.338 usec -> 2021-07-01T02:50:38.230.646.062 (XO: 2.230647270)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 123.463 usec -> 2021-07-01T02:50:38.230.643.187 (XO: 2.230644395)

 - Block number: 2 starting at 2021-07-01T02:51:00.640.788.725 (XO: 24.640791768 - 24.641097769, t=35.6 degC)  Rank: 00078
   TM values: Max =   219  Min =  -257  Ave =    -4.47  Std =    18.98
   Min TM value:-257 at an offset of 128.688 usec -> 2021-07-01T02:51:00.640.917.412 (XO: 24.640920456)
   Max TM value:219 at an offset of 131.400 usec -> 2021-07-01T02:51:00.640.920.125 (XO: 24.640923168)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 131.275 usec -> 2021-07-01T02:51:00.640.920.000 (XO: 24.640923043)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 128.563 usec -> 2021-07-01T02:51:00.640.917.287 (XO: 24.640920331)

 - Block number: 3 starting at 2021-07-01T02:51:11.420.811.725 (XO: 35.420813698 - 35.421119699, t=35.6 degC)  Rank: 00084
   TM values: Max =   223  Min =  -252  Ave =    -4.45  Std =    20.03
   Min TM value:-252 at an offset of 126.563 usec -> 2021-07-01T02:51:11.420.938.287 (XO: 35.420940261)
   Max TM value:223 at an offset of 129.275 usec -> 2021-07-01T02:51:11.420.941.000 (XO: 35.420942973)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 129.175 usec -> 2021-07-01T02:51:11.420.940.900 (XO: 35.420942873)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 126.425 usec -> 2021-07-01T02:51:11.420.938.150 (XO: 35.420940123)

 - Block number: 4 starting at 2021-07-01T02:51:22.031.057.725 (XO: 46.031059034 - 46.031365035, t=35.6 degC)  Rank: 00079
   TM values: Max =   219  Min =  -254  Ave =    -4.49  Std =    22.07
   Min TM value:-254 at an offset of 126.325 usec -> 2021-07-01T02:51:22.031.184.050 (XO: 46.031185359)
   Max TM value:219 at an offset of 126.013 usec -> 2021-07-01T02:51:22.031.183.737 (XO: 46.031185047)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 128.788 usec -> 2021-07-01T02:51:22.031.186.512 (XO: 46.031187822)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 126.113 usec -> 2021-07-01T02:51:22.031.183.837 (XO: 46.031185147)
