-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Aug  1 04:21:48 2021 UTC
-- File 20210630_204751_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.227407 to 39.638205
-each separately from 13s to 39s and from 227407us to 947708us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.719723 in 1 packets: 18
      22.690208 in 1 packets: 36
MIU packet time from 2021-06-30T20:48:03.227.407.000 to 2021-06-30T20:48:29.638.205.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.719723 in 1 packets: 18
      22.690208 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4826051 to 40.4825131
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999894 in 1 packets: 18
      22.9999186 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210630_204751_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.7100133 to 14.7103023
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.7197110  -- OK for SL --
   MIU time from 2021-06-30T20:48:03.227.407.000 to 2021-06-30T20:48:03.227.696.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.719723

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4300133 to 18.4303023
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6901270  -- OK for SL --
   MIU time from 2021-06-30T20:48:06.947.419.000 to 2021-06-30T20:48:06.947.708.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.690208

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.1204293 to 41.1207183
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-30T20:48:29.637.916.000 to 2021-06-30T20:48:29.638.205.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.710013300 corrected by 40bit 10MHz counter of 14.482605100 obtained at 1Hz counter of 13. Result: 13.227409019
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2021-06-30T20:48:03.227.266.725
   Max 1st sample MIU time among all SL packets: 2021-06-30T20:48:29.638.064.725
   Min 1st sample XO time among all SL packets: 13.227268743
   Max 1st sample XO time among all SL packets: 39.638067222
   TM values from -303 to 250 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-30T20:48:03.227.266.725 (XO: 13.227268743 - 13.227574744, t=35.6 degC)  Rank: 00079
   TM values: Max =   230  Min =  -250  Ave =    -4.47  Std =    24.88
   Min TM value:-250 at an offset of 124.913 usec -> 2021-06-30T20:48:03.227.391.637 (XO: 13.227393656)
   Max TM value:230 at an offset of 127.625 usec -> 2021-06-30T20:48:03.227.394.350 (XO: 13.227396369)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 127.475 usec -> 2021-06-30T20:48:03.227.394.200 (XO: 13.227396219)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 124.775 usec -> 2021-06-30T20:48:03.227.391.500 (XO: 13.227393519)

 - Block number: 2 starting at 2021-06-30T20:48:06.947.278.725 (XO: 16.947281935 - 16.947587936, t=35.6 degC)  Rank: 00082
   TM values: Max =   239  Min =  -303  Ave =    -4.34  Std =    27.54
   Min TM value:-303 at an offset of 114.463 usec -> 2021-06-30T20:48:06.947.393.187 (XO: 16.947396398)
   Max TM value:239 at an offset of 117.125 usec -> 2021-06-30T20:48:06.947.395.850 (XO: 16.947399061)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 74.563 usec -> 2021-06-30T20:48:06.947.353.287 (XO: 16.947356498)
   Min dTM/dt [1/us]:-2.36e+03 at an offset of 114.300 usec -> 2021-06-30T20:48:06.947.393.025 (XO: 16.947396236)

 - Block number: 3 starting at 2021-06-30T20:48:29.637.775.725 (XO: 39.637778221 - 39.638084222, t=35.6 degC)  Rank: 00081
   TM values: Max =   250  Min =  -266  Ave =    -4.40  Std =    25.99
   Min TM value:-266 at an offset of 184.138 usec -> 2021-06-30T20:48:29.637.959.863 (XO: 39.637962359)
   Max TM value:250 at an offset of 130.975 usec -> 2021-06-30T20:48:29.637.906.700 (XO: 39.637909196)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 130.900 usec -> 2021-06-30T20:48:29.637.906.625 (XO: 39.637909121)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 183.988 usec -> 2021-06-30T20:48:29.637.959.713 (XO: 39.637962209)
