-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jun  9 13:39:43 2021 UTC
-- File 20210609_132416_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.034535 to 19.923422
-each separately from 13s to 19s and from 34535us to 923422us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.888309 in 1 packets: 18
MIU packet time from 2021-06-09T13:24:32.034.535.000 to 2021-06-09T13:24:38.923.422.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.888309 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8855600 to 20.8855391
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999791 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 8 packets: 2, 3, 4, 5, 6, 7, 8, 9
      2 in 27 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, 20, 21, 22, 23, 24, 25, 26, 27, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210609_132416_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      114 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.9200965 to 14.9203855
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8882870  -- OK for SL --
   MIU time from 2021-06-09T13:24:32.034.535.000 to 2021-06-09T13:24:32.034.824.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.888309

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      25 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.8086725 to 21.8089615
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-06-09T13:24:38.923.133.000 to 2021-06-09T13:24:38.923.422.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.920096500 corrected by 40bit 10MHz counter of 14.885560000 obtained at 1Hz counter of 13. Result: 13.034536621
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-06-09T13:24:32.034.394.725
   Max 1st sample MIU time among all SL packets: 2021-06-09T13:24:38.923.281.725
   Min 1st sample XO time among all SL packets: 13.034396345
   Max 1st sample XO time among all SL packets: 19.923285356
   TM values from -563 to 410 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-06-09T13:24:32.034.394.725 (XO: 13.034396345 - 13.034702346, t=35.6 degC)  Rank: 00114
   TM values: Max =   410  Min =  -563  Ave =    -4.74  Std =    52.03
   Min TM value:-563 at an offset of 105.288 usec -> 2021-06-09T13:24:32.034.500.012 (XO: 13.034501633)
   Max TM value:410 at an offset of 104.088 usec -> 2021-06-09T13:24:32.034.498.812 (XO: 13.034500433)
   Max dTM/dt [1/us]:3.55e+03 at an offset of 98.875 usec -> 2021-06-09T13:24:32.034.493.600 (XO: 13.034495221)
   Min dTM/dt [1/us]:-3.70e+03 at an offset of 279.501 usec -> 2021-06-09T13:24:32.034.674.225 (XO: 13.034675846)

 - Block number: 2 starting at 2021-06-09T13:24:38.922.992.725 (XO: 19.922996355 - 19.923302357, t=35.6 degC)  Rank: 00025
   TM values: Max =    60  Min =  -101  Ave =    -4.52  Std =    13.46
   Min TM value:-101 at an offset of 271.388 usec -> 2021-06-09T13:24:38.923.264.113 (XO: 19.923267744)
   Max TM value:60 at an offset of 271.163 usec -> 2021-06-09T13:24:38.923.263.888 (XO: 19.923267519)
   Max dTM/dt [1/us]:7.94e+02 at an offset of 251.363 usec -> 2021-06-09T13:24:38.923.244.088 (XO: 19.923247719)
   Min dTM/dt [1/us]:-8.94e+02 at an offset of 271.276 usec -> 2021-06-09T13:24:38.923.264.000 (XO: 19.923267631)
