-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20200610_021119_1229.dat, 36720  packets found.
APID:
      1229 (EW) in 36720 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36720 packets (all packets)
Packet no from 1 to 36720
   Differences of packet numbers in neigboring packets: 
      1 in 36719 packets (all packets)
Total number of packets: 
      36720 in 36720 packets (all packets)
Hz and MHz counters from 17.183596 to 45.978495
-each separately from 17s to 45s and from 114089us to 978495us.
   Differences in neigboring packets: 
      0.000000 in 36715 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -22.930493 in 1 packets - #/No: 3672/3672
      22.930493 in 1 packets - #/No: 15912/15912
      5.864406 in 1 packets - #/No: 21216/21216
      -5.864406 in 1 packets - #/No: 33456/33456
MIU packet time from 2020-06-10T02:11:39.183.596.000 to 2020-06-10T02:12:07.978.495.000
   Differences in neigboring packets: 
      0.000000 in 36715 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -22.930493 in 1 packets - #/No: 3672/3672
      22.930493 in 1 packets - #/No: 15912/15912
      5.864406 in 1 packets - #/No: 21216/21216
      -5.864406 in 1 packets - #/No: 33456/33456
MIU orbit number: 
      60000 in 36720 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36720 packets (all packets)
MIU orbit information / number of events: 
      0 in 36720 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36720 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36720 packets (all packets)
FPGA configuration version: 
      211 in 36720 packets (all packets)
10 MHz counter at the last 1Hz pulse from 18.3755983 to 46.3755040
   Differences in neigboring packets: 
      0.0000000 in 36715 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -22.9999226 in 1 packets - #/No: 3672/3672
      22.9999226 in 1 packets - #/No: 15912/15912
      4.9999831 in 1 packets - #/No: 21216/21216
      -4.9999831 in 1 packets - #/No: 33456/33456
Temperature FPGA: 
      06 -> 47.2°C in 36720 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36720 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 36720 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 36720 packets (all packets)
Number of external events: 
      2 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 12240 packets - #/No: 3673/3673, 3674/3674, 3675/3675, 3676/3676, 3677/3677, 3678/3678, 3679/3679, 3680/3680, 3681/3681, 3682/3682, 3683/3683, 3684/3684, 3685/3685, 3686/3686, 3687/3687, 3688/3688, 3689/3689, 3690/3690, ...
      3 in 12240 packets - #/No: 21217/21217, 21218/21218, 21219/21219, 21220/21220, 21221/21221, 21222/21222, 21223/21223, 21224/21224, 21225/21225, 21226/21226, 21227/21227, 21228/21228, 21229/21229, 21230/21230, 21231/21231, 21232/21232, 21233/21233, 21234/21234, ...
Number of generated alerts: 
      3 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 12240 packets - #/No: 3673/3673, 3674/3674, 3675/3675, 3676/3676, 3677/3677, 3678/3678, 3679/3679, 3680/3680, 3681/3681, 3682/3682, 3683/3683, 3684/3684, 3685/3685, 3686/3686, 3687/3687, 3688/3688, 3689/3689, 3690/3690, ...
      12 in 12240 packets - #/No: 21217/21217, 21218/21218, 21219/21219, 21220/21220, 21221/21221, 21222/21222, 21223/21223, 21224/21224, 21225/21225, 21226/21226, 21227/21227, 21228/21228, 21229/21229, 21230/21230, 21231/21231, 21232/21232, 21233/21233, 21234/21234, ...
Survey memory error: 
      0 in 36720 packets (all packets)
Event memory error: 
      0 in 36720 packets (all packets)
Configuration error: 
      0 in 36720 packets (all packets)
Zero: 
      0 in 36720 packets (all packets)
Radio gain: 
      0 in 36720 packets (all packets)
Radio bandwidth: 
      0 in 36720 packets (all packets)
Digital gain: 
      0 in 36720 packets (all packets)
Analog gain: 
      0 in 36720 packets (all packets)

EVENT DATA READER
-- File 20200610_021119_1229.dat, 180 EW pages found. Extra packets: 0.
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      2 in 12240 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 12240 packets - #/No: 3673/3673, 3674/3674, 3675/3675, 3676/3676, 3677/3677, 3678/3678, 3679/3679, 3680/3680, 3681/3681, 3682/3682, 3683/3683, 3684/3684, 3685/3685, 3686/3686, 3687/3687, 3688/3688, 3689/3689, 3690/3690, ...
      3 in 12240 packets - #/No: 21217/21217, 21218/21218, 21219/21219, 21220/21220, 21221/21221, 21222/21222, 21223/21223, 21224/21224, 21225/21225, 21226/21226, 21227/21227, 21228/21228, 21229/21229, 21230/21230, 21231/21231, 21232/21232, 21233/21233, 21234/21234, ...
   10 MHz counter from 18.5037069 to 47.5065739
      Difference between the neighboring packets: 
      0.0000170 in 36714 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      -23.0483110 in 1 packets - #/No: 3672/3672
      22.8402650 in 1 packets - #/No: 15912/15912
      5.7291530 in 1 packets - #/No: 21216/21216
      -0.2080630 in 1 packets - #/No: 29172/29172
      -5.9371990 in 1 packets - #/No: 33456/33456
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 178 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3468/3468, 3672/3672, ...
      59 in 1 packets - #/No: 29172/29172
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 36540 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      1 in 174 packets - #/No: 204/204, 408/408, 612/612, 816/816, 1020/1020, 1224/1224, 1428/1428, 1632/1632, 1836/1836, 2040/2040, 2244/2244, 2448/2448, 2652/2652, 2856/2856, 3060/3060, 3264/3264, 3468/3468, 3876/3876, ...
      -33 in 1 packets - #/No: 3672/3672
      -25 in 1 packets - #/No: 15912/15912
      -38 in 1 packets - #/No: 21216/21216
      -59 in 1 packets - #/No: 29172/29172
      -20 in 1 packets - #/No: 33456/33456
   MIU event flag 
      1 in 36720 packets (all packets)
   Time triggered event flag 
      0 in 36720 packets (all packets)
   Min EW event (1st sample) abs. time: 2020-06-10T02:11:39.128.107.562
   Min EW header abs. time            : 2020-06-10T02:11:39.183.596.000
   Max EW event (1st sample) abs. time: 2020-06-10T02:12:07.923.006.567
   Max EW header abs. time            : 2020-06-10T02:12:07.978.495.000
   TM values from -2048 to 1579 median:-5.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -764 to 420 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=17 C1MHz=0.183596
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=18.5037069
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2020-06-10T02:11:39.128.107.562
      (header abs. time: 2020-06-10T02:11:39.183.596.000)
      Event 1st sample XO time (corr. sec. from XP start):              17.128108786
      Event header XO time (corr. sec. from XP start)    :              17.128109036
      Sampling frequency correction factor for this event:     0.999996600

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -2048 to 1579 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=40 C1MHz=0.114089
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=41.4341229
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2020-06-10T02:12:02.058.600.562
      (header abs. time: 2020-06-10T02:12:02.114.089.000)
      Event 1st sample XO time (corr. sec. from XP start):              40.058601949
      Event header XO time (corr. sec. from XP start)    :              40.058602199
      Sampling frequency correction factor for this event:     0.999996600

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      216 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      217 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      218 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      219 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      220 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      113 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      114 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      115 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      116 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      117 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      123 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      124 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      125 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      126 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      127 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      128 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      129 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      130 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      131 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      132 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      133 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      134 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      135 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      136 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      137 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      138 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      139 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      140 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      141 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      142 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -719 to 1246 median:-5.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=45 C1MHz=0.978495
   Differences of 10-MHz counter in neigboring packets of this event (#/No=packet/page): 
      0.0000170 in 12239 packets (all packets)
   10MHz counter -- OK -- (correct differences for all packets of this event: EM4 design):
          First C10MHz=47.2985109
      Event type -- OK -- (constant for all packets of this event): External
      Gain -- OK -- (constant for all packets of this event): 1x
      Event 1st sample abs. time -- OK -- (constant for all packets of this event): 2020-06-10T02:12:07.923.006.567
      (header abs. time: 2020-06-10T02:12:07.978.495.000)
      Event 1st sample XO time (corr. sec. from XP start):              45.923006696
      Event header XO time (corr. sec. from XP start)    :              45.923006946
      Sampling frequency correction factor for this event:     0.999996700
--
Extremes in separate events:

 - Event number: 1 starting at 2020-06-10T02:11:39.128.107.562 (XO: 17.1281088)
   Min TM value:-764 at an offset of 54129.134 usec -> 2020-06-10T02:11:39.182.236.696 (XO: 17.1822379)
   Max TM value:420 at an offset of 55699.302 usec -> 2020-06-10T02:11:39.183.806.863 (XO: 17.1838081)
   Max dTM/dt [1/us]:3.86e+03 at an offset of 52992.218 usec -> 2020-06-10T02:11:39.181.099.779 (XO: 17.1811010)
   Min dTM/dt [1/us]:-7.11e+03 at an offset of 54129.097 usec -> 2020-06-10T02:11:39.182.236.658 (XO: 17.1822379)

 - Event number: 2 starting at 2020-06-10T02:12:02.058.600.562 (XO: 40.0586019)
   Min TM value:-2048 at an offset of 56476.980 usec -> 2020-06-10T02:12:02.115.077.541 (XO: 40.1150789)
   Max TM value:1579 at an offset of 56477.992 usec -> 2020-06-10T02:12:02.115.078.554 (XO: 40.1150799)
   Max dTM/dt [1/us]:1.51e+04 at an offset of 56477.280 usec -> 2020-06-10T02:12:02.115.077.841 (XO: 40.1150792)
   Min dTM/dt [1/us]:-1.21e+04 at an offset of 56476.192 usec -> 2020-06-10T02:12:02.115.076.754 (XO: 40.1150781)

 - Event number: 3 starting at 2020-06-10T02:12:07.923.006.567 (XO: 45.9230067)
   Min TM value:-719 at an offset of 54173.391 usec -> 2020-06-10T02:12:07.977.179.958 (XO: 45.9771801)
   Max TM value:1246 at an offset of 63724.473 usec -> 2020-06-10T02:12:07.986.731.039 (XO: 45.9867312)
   Max dTM/dt [1/us]:1.11e+04 at an offset of 63724.435 usec -> 2020-06-10T02:12:07.986.731.002 (XO: 45.9867311)
   Min dTM/dt [1/us]:-1.41e+04 at an offset of 63724.523 usec -> 2020-06-10T02:12:07.986.731.089 (XO: 45.9867312)
