-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180713_005243_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.296112 to 31.739026
-each separately from 0s to 31s and from 296112us to 739026us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.078818 in 1 packets - #/No: 18/18
      3.363229 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-13T00:52:45.296.112.000 to 2018-07-13T00:53:16.739.026.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      28.078818 in 1 packets - #/No: 18/18
      3.363229 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4347485 to 32.4346210
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      27.9998849 in 1 packets - #/No: 18/18
      2.9999876 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180713_005243_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7308602 to 1.7311492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.0787030  -- OK for SL --
   MIU time from 2018-07-13T00:52:45.296.112.000 to 2018-07-13T00:52:45.296.401.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.078818

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8098522 to 29.8101412
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.3632150  -- OK for SL --
   MIU time from 2018-07-13T00:53:13.375.219.000 to 2018-07-13T00:53:13.375.508.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.363229

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.1733562 to 33.1736452
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-13T00:53:16.738.737.000 to 2018-07-13T00:53:16.739.026.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-13T00:52:45.295.971.750
   Max SL corrected packet time: 2018-07-13T00:53:16.738.885.750
   Min SL corrected XO rel time: 0.2961129
   Max SL corrected XO rel time: 31.7390272
   TM values from -21 to 13 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-13T00:52:45.295.971.750 (XO: 0.2961129, t=34.0°C)  Rank: 00004
   TM values: Max =    10  Min =   -21  Ave =    -4.26  Std =     5.56
   Min TM value:-21 at an offset of 130.201 usec -> 2018-07-13T00:52:45.296.101.950 (XO: 0.2962431)
   Max TM value:10 at an offset of 31.413 usec -> 2018-07-13T00:52:45.296.003.162 (XO: 0.2961443)
   Max dTM/dt [1/us]::2.24e+02 at an offset of 230.551 usec -> 2018-07-13T00:52:45.296.202.300 (XO: 0.2963435)
   Min dTM/dt [1/us]::-2.09e+02 at an offset of 271.676 usec -> 2018-07-13T00:52:45.296.243.426 (XO: 0.2963846)

 - Block number: 2 starting at 2018-07-13T00:53:13.375.078.750 (XO: 28.3752204, t=34.0°C)  Rank: 00006
   TM values: Max =    13  Min =   -21  Ave =    -4.37  Std =     4.97
   Min TM value:-21 at an offset of 107.300 usec -> 2018-07-13T00:53:13.375.186.050 (XO: 28.3753277)
   Max TM value:13 at an offset of 171.188 usec -> 2018-07-13T00:53:13.375.249.938 (XO: 28.3753916)
   Max dTM/dt [1/us]::2.17e+02 at an offset of 117.988 usec -> 2018-07-13T00:53:13.375.196.737 (XO: 28.3753384)
   Min dTM/dt [1/us]::-2.19e+02 at an offset of 108.113 usec -> 2018-07-13T00:53:13.375.186.862 (XO: 28.3753285)

 - Block number: 3 starting at 2018-07-13T00:53:16.738.596.750 (XO: 31.7387382, t=34.0°C)  Rank: 00005
   TM values: Max =    11  Min =   -21  Ave =    -4.32  Std =     5.07
   Min TM value:-21 at an offset of 125.076 usec -> 2018-07-13T00:53:16.738.721.825 (XO: 31.7388633)
   Max TM value:11 at an offset of 71.313 usec -> 2018-07-13T00:53:16.738.668.062 (XO: 31.7388095)
   Max dTM/dt [1/us]::2.13e+02 at an offset of 74.113 usec -> 2018-07-13T00:53:16.738.670.862 (XO: 31.7388123)
   Min dTM/dt [1/us]::-2.33e+02 at an offset of 125.051 usec -> 2018-07-13T00:53:16.738.721.800 (XO: 31.7388633)
