-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180709_173309_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 11.437670 to 35.503580
-each separately from 11s to 35s and from 437670us to 698433us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.260185 in 1 packets - #/No: 18/18
      17.804858 in 1 packets - #/No: 36/18
MIU packet time from 2018-07-09T17:33:23.437.670.000 to 2018-07-09T17:33:47.503.580.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.260185 in 1 packets - #/No: 18/18
      17.804858 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.4378109 to 36.4377100
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.9999747 in 1 packets - #/No: 18/18
      17.9999244 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180709_173309_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.8754805 to 12.8757695
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.2601590  -- OK for SL --
   MIU time from 2018-07-09T17:33:23.437.670.000 to 2018-07-09T17:33:23.437.959.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.260185

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.1359285 to 19.1362175
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8047830  -- OK for SL --
   MIU time from 2018-07-09T17:33:29.698.144.000 to 2018-07-09T17:33:29.698.433.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.804858

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.9410005 to 36.9412895
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-07-09T17:33:47.503.291.000 to 2018-07-09T17:33:47.503.580.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-07-09T17:33:23.437.529.750
   Max SL corrected packet time: 2018-07-09T17:33:47.503.439.750
   Min SL corrected XO rel time: 11.4376714
   Max SL corrected XO rel time: 35.5035816
   TM values from -35 to 27 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-07-09T17:33:23.437.529.750 (XO: 11.4376714, t=34.0°C)  Rank: 00006
   TM values: Max =    20  Min =   -30  Ave =    -4.41  Std =     6.78
   Min TM value:-30 at an offset of 123.651 usec -> 2018-07-09T17:33:23.437.653.400 (XO: 11.4377951)
   Max TM value:20 at an offset of 96.663 usec -> 2018-07-09T17:33:23.437.626.412 (XO: 11.4377681)
   Max dTM/dt [1/us]::2.55e+02 at an offset of 123.176 usec -> 2018-07-09T17:33:23.437.652.925 (XO: 11.4377946)
   Min dTM/dt [1/us]::-2.18e+02 at an offset of 126.501 usec -> 2018-07-09T17:33:23.437.656.250 (XO: 11.4377979)

 - Block number: 2 starting at 2018-07-09T17:33:29.698.003.750 (XO: 17.6981458, t=34.0°C)  Rank: 00006
   TM values: Max =    24  Min =   -33  Ave =    -4.39  Std =     7.04
   Min TM value:-33 at an offset of 113.625 usec -> 2018-07-09T17:33:29.698.117.375 (XO: 17.6982594)
   Max TM value:24 at an offset of 143.513 usec -> 2018-07-09T17:33:29.698.147.263 (XO: 17.6982893)
   Max dTM/dt [1/us]::2.71e+02 at an offset of 113.800 usec -> 2018-07-09T17:33:29.698.117.550 (XO: 17.6982596)
   Min dTM/dt [1/us]::-2.64e+02 at an offset of 83.913 usec -> 2018-07-09T17:33:29.698.087.662 (XO: 17.6982297)

 - Block number: 3 starting at 2018-07-09T17:33:47.503.150.750 (XO: 35.5032926, t=34.0°C)  Rank: 00006
   TM values: Max =    27  Min =   -35  Ave =    -4.30  Std =     8.79
   Min TM value:-35 at an offset of 130.301 usec -> 2018-07-09T17:33:47.503.281.050 (XO: 35.5034229)
   Max TM value:27 at an offset of 197.838 usec -> 2018-07-09T17:33:47.503.348.588 (XO: 35.5034905)
   Max dTM/dt [1/us]::3.00e+02 at an offset of 252.014 usec -> 2018-07-09T17:33:47.503.402.763 (XO: 35.5035446)
   Min dTM/dt [1/us]::-3.01e+02 at an offset of 144.763 usec -> 2018-07-09T17:33:47.503.295.513 (XO: 35.5034374)
