-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171007_122504_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.241191 to 30.024637
-each separately from 0s to 30s and from 24348us to 709097us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.467328 in 1 packets - #/No: 18/18
      13.315251 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-07T12:25:07.241.191.000 to 2017-10-07T12:25:37.024.637.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.467328 in 1 packets - #/No: 18/18
      13.315251 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9402632 to 31.9401457
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999375 in 1 packets - #/No: 18/18
      13.9999450 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171007_122504_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1814549 to 2.1817439
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4672630  -- OK for SL --
   MIU time from 2017-10-07T12:25:07.241.191.000 to 2017-10-07T12:25:07.241.480.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.467328

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.6490069 to 18.6492959
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3151990  -- OK for SL --
   MIU time from 2017-10-07T12:25:23.708.808.000 to 2017-10-07T12:25:23.709.097.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.315251

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9644949 to 31.9647839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-07T12:25:37.024.348.000 to 2017-10-07T12:25:37.024.637.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-07T12:25:07.241.050.750
   Max SL corrected packet time: 2017-10-07T12:25:37.024.496.750
   Min SL corrected XO rel time: 0.2411926
   Max SL corrected XO rel time: 30.0246383
   TM values from -18 to 10 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-07T12:25:07.241.050.750 (XO: 0.2411926, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -17  Ave =    -4.45  Std =     3.65
   Min TM value:-17 at an offset of 84.738 usec -> 2017-10-07T12:25:07.241.135.487 (XO: 0.2412774)
   Max TM value:9 at an offset of 94.763 usec -> 2017-10-07T12:25:07.241.145.512 (XO: 0.2412874)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 105.063 usec -> 2017-10-07T12:25:07.241.155.812 (XO: 0.2412977)
   Min dTM/dt [1/us]::-9.70e+01 at an offset of 127.000 usec -> 2017-10-07T12:25:07.241.177.750 (XO: 0.2413196)

 - Block number: 2 starting at 2017-10-07T12:25:23.708.667.750 (XO: 16.7088090, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -18  Ave =    -4.47  Std =     3.71
   Min TM value:-18 at an offset of 89.500 usec -> 2017-10-07T12:25:23.708.757.250 (XO: 16.7088985)
   Max TM value:10 at an offset of 110.025 usec -> 2017-10-07T12:25:23.708.777.775 (XO: 16.7089191)
   Max dTM/dt [1/us]::1.08e+02 at an offset of 109.650 usec -> 2017-10-07T12:25:23.708.777.400 (XO: 16.7089187)
   Min dTM/dt [1/us]::-1.08e+02 at an offset of 110.075 usec -> 2017-10-07T12:25:23.708.777.825 (XO: 16.7089191)

 - Block number: 3 starting at 2017-10-07T12:25:37.024.207.750 (XO: 30.0243493, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -18  Ave =    -4.36  Std =     4.30
   Min TM value:-18 at an offset of 88.363 usec -> 2017-10-07T12:25:37.024.296.112 (XO: 30.0244377)
   Max TM value:10 at an offset of 79.750 usec -> 2017-10-07T12:25:37.024.287.500 (XO: 30.0244290)
   Max dTM/dt [1/us]::1.03e+02 at an offset of 144.263 usec -> 2017-10-07T12:25:37.024.352.013 (XO: 30.0244936)
   Min dTM/dt [1/us]::-1.01e+02 at an offset of 95.213 usec -> 2017-10-07T12:25:37.024.302.962 (XO: 30.0244445)
